-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 03:56:17 2021 UTC
-- File 20210727_105751_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.531987 to 18.144443
-each separately from 11s to 18s and from 144154us to 532276us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.611878 in 1 packets: 18
MIU packet time from 2021-07-27T10:58:05.531.987.000 to 2021-07-27T10:58:12.144.443.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.611878 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8747257 to 19.8747021
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999764 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210727_105751_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4067129 to 13.4070019
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6118550  -- OK for SL --
   MIU time from 2021-07-27T10:58:05.531.987.000 to 2021-07-27T10:58:05.532.276.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.611878

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0188569 to 20.0191459
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-27T10:58:12.144.154.000 to 2021-07-27T10:58:12.144.443.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.406712900 corrected by 40bit 10MHz counter of 12.874725700 obtained at 1Hz counter of 11. Result: 11.531989009
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-27T10:58:05.531.846.725
   Max 1st sample MIU time among all SL packets: 2021-07-27T10:58:12.144.302.725
   Min 1st sample XO time among all SL packets: 11.531848733
   Max 1st sample XO time among all SL packets: 18.144304030
   TM values from -285 to 261 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-27T10:58:05.531.846.725 (XO: 11.531848733 - 11.532154734, t=37.2 degC)  Rank: 00083
   TM values: Max =   261  Min =  -285  Ave =    -4.56  Std =    46.71
   Min TM value:-285 at an offset of 182.888 usec -> 2021-07-27T10:58:05.532.029.613 (XO: 11.532031621)
   Max TM value:261 at an offset of 175.163 usec -> 2021-07-27T10:58:05.532.021.888 (XO: 11.532023896)
   Max dTM/dt [1/us]:2.62e+03 at an offset of 194.238 usec -> 2021-07-27T10:58:05.532.040.963 (XO: 11.532042971)
   Min dTM/dt [1/us]:-2.96e+03 at an offset of 225.363 usec -> 2021-07-27T10:58:05.532.072.088 (XO: 11.532074097)

 - Block number: 2 starting at 2021-07-27T10:58:12.144.013.725 (XO: 18.144015029 - 18.144321030, t=37.2 degC)  Rank: 00082
   TM values: Max =   242  Min =  -247  Ave =    -4.44  Std =    18.13
   Min TM value:-247 at an offset of 110.375 usec -> 2021-07-27T10:58:12.144.124.100 (XO: 18.144125404)
   Max TM value:242 at an offset of 110.125 usec -> 2021-07-27T10:58:12.144.123.850 (XO: 18.144125154)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 112.788 usec -> 2021-07-27T10:58:12.144.126.512 (XO: 18.144127817)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 110.225 usec -> 2021-07-27T10:58:12.144.123.950 (XO: 18.144125254)
