-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 29 00:31:32 2021 UTC
-- File 20210727_090430_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.663205 to 45.667299
-each separately from 12s to 45s and from 508267us to 810977us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.147194 in 1 packets: 18
      12.697290 in 1 packets: 36
      7.158454 in 1 packets: 54
MIU packet time from 2021-07-27T09:04:45.663.205.000 to 2021-07-27T09:05:18.667.299.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.147194 in 1 packets: 18
      12.697290 in 1 packets: 36
      7.158454 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8434202 to 46.8433108
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999569 in 2 packets: 18, 36
      6.9999768 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210727_090430_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.5066256 to 14.5069146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1471510  -- OK for SL --
   MIU time from 2021-07-27T09:04:45.663.205.000 to 2021-07-27T09:04:45.663.494.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.147194

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.6540656 to 27.6543546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6972470  -- OK for SL --
   MIU time from 2021-07-27T09:04:58.810.688.000 to 2021-07-27T09:04:58.810.977.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.697290

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.3516016 to 40.3518906
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.1584310  -- OK for SL --
   MIU time from 2021-07-27T09:05:11.508.267.000 to 2021-07-27T09:05:11.508.556.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.158454

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.5103216 to 47.5106106
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-27T09:05:18.667.010.000 to 2021-07-27T09:05:18.667.299.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.506625600 corrected by 40bit 10MHz counter of 13.843420200 obtained at 1Hz counter of 12. Result: 12.663207589
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-27T09:04:45.663.064.725
   Max 1st sample MIU time among all SL packets: 2021-07-27T09:05:18.667.158.725
   Min 1st sample XO time among all SL packets: 12.663067313
   Max 1st sample XO time among all SL packets: 45.667161727
   TM values from -236 to 210 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-27T09:04:45.663.064.725 (XO: 12.663067313 - 12.663373314, t=37.2 degC)  Rank: 00084
   TM values: Max =   196  Min =  -236  Ave =    -4.52  Std =    18.71
   Min TM value:-236 at an offset of 128.063 usec -> 2021-07-27T09:04:45.663.192.787 (XO: 12.663195376)
   Max TM value:196 at an offset of 130.688 usec -> 2021-07-27T09:04:45.663.195.412 (XO: 12.663198001)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 130.538 usec -> 2021-07-27T09:04:45.663.195.262 (XO: 12.663197851)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 127.913 usec -> 2021-07-27T09:04:45.663.192.637 (XO: 12.663195226)

 - Block number: 2 starting at 2021-07-27T09:04:58.810.547.725 (XO: 25.810550900 - 25.810856901, t=37.2 degC)  Rank: 00081
   TM values: Max =   192  Min =  -236  Ave =    -4.51  Std =    20.99
   Min TM value:-236 at an offset of 124.825 usec -> 2021-07-27T09:04:58.810.672.550 (XO: 25.810675725)
   Max TM value:192 at an offset of 124.525 usec -> 2021-07-27T09:04:58.810.672.250 (XO: 25.810675425)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 127.225 usec -> 2021-07-27T09:04:58.810.674.950 (XO: 25.810678125)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 124.613 usec -> 2021-07-27T09:04:58.810.672.337 (XO: 25.810675513)

 - Block number: 3 starting at 2021-07-27T09:05:11.508.126.725 (XO: 38.508129002 - 38.508435003, t=37.2 degC)  Rank: 00079
   TM values: Max =   185  Min =  -229  Ave =    -4.57  Std =    17.65
   Min TM value:-229 at an offset of 126.438 usec -> 2021-07-27T09:05:11.508.253.162 (XO: 38.508255440)
   Max TM value:185 at an offset of 129.113 usec -> 2021-07-27T09:05:11.508.255.837 (XO: 38.508258115)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 128.938 usec -> 2021-07-27T09:05:11.508.255.662 (XO: 38.508257940)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 126.225 usec -> 2021-07-27T09:05:11.508.252.950 (XO: 38.508255227)

 - Block number: 4 starting at 2021-07-27T09:05:18.666.869.725 (XO: 45.666872726 - 45.667178727, t=37.2 degC)  Rank: 00079
   TM values: Max =   210  Min =  -232  Ave =    -4.55  Std =    26.40
   Min TM value:-232 at an offset of 158.338 usec -> 2021-07-27T09:05:18.667.028.063 (XO: 45.667031064)
   Max TM value:210 at an offset of 115.588 usec -> 2021-07-27T09:05:18.666.985.312 (XO: 45.666988314)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 118.263 usec -> 2021-07-27T09:05:18.666.987.987 (XO: 45.666990989)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 115.688 usec -> 2021-07-27T09:05:18.666.985.412 (XO: 45.666988414)
