-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 28 21:51:40 2021 UTC
-- File 20210727_032911_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.815020 to 41.228728
-each separately from 9s to 41s and from 228439us to 815309us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.820085 in 1 packets: 18
      15.592756 in 1 packets: 36
MIU packet time from 2021-07-27T03:29:23.815.020.000 to 2021-07-27T03:29:55.228.728.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.820085 in 1 packets: 18
      15.592756 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9523719 to 42.9522646
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999464 in 1 packets: 18
      15.9999463 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 3 packets: 37, 38, 39
      4 in 12 packets: 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51
      5 in 3 packets: 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210727_032911_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7673927 to 11.7676817
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8200310  -- OK for SL --
   MIU time from 2021-07-27T03:29:23.815.020.000 to 2021-07-27T03:29:23.815.309.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.820085

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.5877127 to 27.5880017
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5927030  -- OK for SL --
   MIU time from 2021-07-27T03:29:39.635.394.000 to 2021-07-27T03:29:39.635.683.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.592756

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      169 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1807047 to 43.1809937
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-27T03:29:55.228.439.000 to 2021-07-27T03:29:55.228.728.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.767392700 corrected by 40bit 10MHz counter of 10.952371900 obtained at 1Hz counter of 9. Result: 9.815023571
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-27T03:29:23.814.879.725
   Max 1st sample MIU time among all SL packets: 2021-07-27T03:29:55.228.587.725
   Min 1st sample XO time among all SL packets: 9.814883296
   Max 1st sample XO time among all SL packets: 41.228589579
   TM values from -795 to 471 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-27T03:29:23.814.879.725 (XO: 9.814883296 - 9.815189297, t=37.2 degC)  Rank: 00077
   TM values: Max =   205  Min =  -239  Ave =    -4.45  Std =    22.31
   Min TM value:-239 at an offset of 120.925 usec -> 2021-07-27T03:29:23.815.000.650 (XO: 9.815004221)
   Max TM value:205 at an offset of 120.625 usec -> 2021-07-27T03:29:23.815.000.350 (XO: 9.815003921)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 123.288 usec -> 2021-07-27T03:29:23.815.003.012 (XO: 9.815006584)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 120.725 usec -> 2021-07-27T03:29:23.815.000.450 (XO: 9.815004021)

 - Block number: 2 starting at 2021-07-27T03:29:39.635.253.725 (XO: 25.635256285 - 25.635562286, t=37.2 degC)  Rank: 00077
   TM values: Max =   169  Min =  -219  Ave =    -4.42  Std =    19.16
   Min TM value:-219 at an offset of 117.763 usec -> 2021-07-27T03:29:39.635.371.487 (XO: 25.635374048)
   Max TM value:169 at an offset of 120.450 usec -> 2021-07-27T03:29:39.635.374.175 (XO: 25.635376735)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 120.250 usec -> 2021-07-27T03:29:39.635.373.975 (XO: 25.635376535)
   Min dTM/dt [1/us]:-1.52e+03 at an offset of 88.700 usec -> 2021-07-27T03:29:39.635.342.425 (XO: 25.635344985)

 - Block number: 3 starting at 2021-07-27T03:29:55.228.298.725 (XO: 41.228300578 - 41.228606579, t=37.2 degC)  Rank: 00169
   TM values: Max =   471  Min =  -795  Ave =    -4.37  Std =    66.85
   Min TM value:-795 at an offset of 134.050 usec -> 2021-07-27T03:29:55.228.432.775 (XO: 41.228434629)
   Max TM value:471 at an offset of 136.400 usec -> 2021-07-27T03:29:55.228.435.125 (XO: 41.228436979)
   Max dTM/dt [1/us]:6.57e+03 at an offset of 134.125 usec -> 2021-07-27T03:29:55.228.432.850 (XO: 41.228434704)
   Min dTM/dt [1/us]:-4.80e+03 at an offset of 125.613 usec -> 2021-07-27T03:29:55.228.424.337 (XO: 41.228426191)
