-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 28 14:56:11 2021 UTC
-- File 20210726_161857_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.515581 to 27.775909
-each separately from 6s to 27s and from 515581us to 775909us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.259750 in 1 packets: 18
MIU packet time from 2021-07-26T16:19:06.515.581.000 to 2021-07-26T16:19:27.775.909.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.259750 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8211316 to 28.8210606
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999290 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210726_161857_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3367134 to 8.3370024
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2596790  -- OK for SL --
   MIU time from 2021-07-26T16:19:06.515.581.000 to 2021-07-26T16:19:06.515.870.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.259750

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5966814 to 29.5969704
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-26T16:19:27.775.620.000 to 2021-07-26T16:19:27.775.909.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.336713400 corrected by 40bit 10MHz counter of 7.821131600 obtained at 1Hz counter of 6. Result: 6.515583553
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-26T16:19:06.515.440.725
   Max 1st sample MIU time among all SL packets: 2021-07-26T16:19:27.775.768.725
   Min 1st sample XO time among all SL packets: 6.515443278
   Max 1st sample XO time among all SL packets: 27.775772085
   TM values from -280 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-26T16:19:06.515.440.725 (XO: 6.515443278 - 6.515749279, t=37.2 degC)  Rank: 00073
   TM values: Max =   231  Min =  -280  Ave =    -4.44  Std =    23.57
   Min TM value:-280 at an offset of 36.338 usec -> 2021-07-26T16:19:06.515.477.062 (XO: 6.515479615)
   Max TM value:231 at an offset of 36.638 usec -> 2021-07-26T16:19:06.515.477.362 (XO: 6.515479915)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 36.450 usec -> 2021-07-26T16:19:06.515.477.175 (XO: 6.515479728)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 33.875 usec -> 2021-07-26T16:19:06.515.474.600 (XO: 6.515477153)

 - Block number: 2 starting at 2021-07-26T16:19:27.775.479.725 (XO: 27.775483084 - 27.775789085, t=37.2 degC)  Rank: 00079
   TM values: Max =   235  Min =  -280  Ave =    -4.50  Std =    23.94
   Min TM value:-280 at an offset of 107.263 usec -> 2021-07-26T16:19:27.775.586.987 (XO: 27.775590347)
   Max TM value:235 at an offset of 122.513 usec -> 2021-07-26T16:19:27.775.602.237 (XO: 27.775605597)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 122.400 usec -> 2021-07-26T16:19:27.775.602.125 (XO: 27.775605484)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 119.650 usec -> 2021-07-26T16:19:27.775.599.375 (XO: 27.775602734)
