-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 28 14:43:19 2021 UTC
-- File 20210726_161317_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.794660 to 17.654840
-each separately from 2s to 17s and from 654551us to 794949us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.859602 in 1 packets: 18
MIU packet time from 2021-07-26T16:13:22.794.660.000 to 2021-07-26T16:13:37.654.840.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.859602 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9009503 to 18.9008997
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999494 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210726_161317_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6956112 to 4.6959002
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8595510  -- OK for SL --
   MIU time from 2021-07-26T16:13:22.794.660.000 to 2021-07-26T16:13:22.794.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.859602

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5554512 to 19.5557402
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-26T16:13:37.654.551.000 to 2021-07-26T16:13:37.654.840.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.695611200 corrected by 40bit 10MHz counter of 3.900950300 obtained at 1Hz counter of 2. Result: 2.794663522
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-26T16:13:22.794.519.725
   Max 1st sample MIU time among all SL packets: 2021-07-26T16:13:37.654.699.725
   Min 1st sample XO time among all SL packets: 2.794523247
   Max 1st sample XO time among all SL packets: 17.654702451
   TM values from -268 to 222 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-26T16:13:22.794.519.725 (XO: 2.794523247 - 2.794829248, t=37.2 degC)  Rank: 00078
   TM values: Max =   214  Min =  -244  Ave =    -4.33  Std =    23.67
   Min TM value:-244 at an offset of 165.713 usec -> 2021-07-26T16:13:22.794.685.438 (XO: 2.794688960)
   Max TM value:214 at an offset of 123.125 usec -> 2021-07-26T16:13:22.794.642.850 (XO: 2.794646372)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 122.988 usec -> 2021-07-26T16:13:22.794.642.712 (XO: 2.794646235)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 108.175 usec -> 2021-07-26T16:13:22.794.627.900 (XO: 2.794631422)

 - Block number: 2 starting at 2021-07-26T16:13:37.654.410.725 (XO: 17.654413450 - 17.654719451, t=37.2 degC)  Rank: 00084
   TM values: Max =   222  Min =  -268  Ave =    -4.35  Std =    22.53
   Min TM value:-268 at an offset of 94.850 usec -> 2021-07-26T16:13:37.654.505.575 (XO: 17.654508300)
   Max TM value:222 at an offset of 110.038 usec -> 2021-07-26T16:13:37.654.520.762 (XO: 17.654523488)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 109.825 usec -> 2021-07-26T16:13:37.654.520.550 (XO: 17.654523275)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 107.250 usec -> 2021-07-26T16:13:37.654.517.975 (XO: 17.654520700)
