-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 26 19:16:58 2021 UTC
-- File 20210726_094623_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.476323 to 38.455522
-each separately from 10s to 38s and from 455233us to 486267us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.009366 in 1 packets: 18
      20.968966 in 1 packets: 36
MIU packet time from 2021-07-26T09:46:36.476.323.000 to 2021-07-26T09:47:04.455.522.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.009366 in 1 packets: 18
      20.968966 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8496402 to 39.8495462
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999765 in 1 packets: 18
      20.9999295 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210726_094623_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3259636 to 12.3262526
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.0093430  -- OK for SL --
   MIU time from 2021-07-26T09:46:36.476.323.000 to 2021-07-26T09:46:36.476.612.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.009366

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3355956 to 19.3358846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.9688950  -- OK for SL --
   MIU time from 2021-07-26T09:46:43.485.978.000 to 2021-07-26T09:46:43.486.267.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.968966

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.3047796 to 40.3050686
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-26T09:47:04.455.233.000 to 2021-07-26T09:47:04.455.522.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.325963600 corrected by 40bit 10MHz counter of 11.849640200 obtained at 1Hz counter of 10. Result: 10.476325020
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-26T09:46:36.476.182.725
   Max 1st sample MIU time among all SL packets: 2021-07-26T09:47:04.455.381.725
   Min 1st sample XO time among all SL packets: 10.476184744
   Max 1st sample XO time among all SL packets: 38.455383673
   TM values from -256 to 240 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-26T09:46:36.476.182.725 (XO: 10.476184744 - 10.476490745, t=37.2 degC)  Rank: 00076
   TM values: Max =   182  Min =  -245  Ave =    -4.44  Std =    23.49
   Min TM value:-245 at an offset of 124.075 usec -> 2021-07-26T09:46:36.476.306.800 (XO: 10.476308819)
   Max TM value:182 at an offset of 126.663 usec -> 2021-07-26T09:46:36.476.309.387 (XO: 10.476311407)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 126.488 usec -> 2021-07-26T09:46:36.476.309.212 (XO: 10.476311232)
   Min dTM/dt [1/us]:-1.52e+03 at an offset of 123.850 usec -> 2021-07-26T09:46:36.476.306.575 (XO: 10.476308594)

 - Block number: 2 starting at 2021-07-26T09:46:43.485.837.725 (XO: 17.485840277 - 17.486146278, t=37.2 degC)  Rank: 00088
   TM values: Max =   240  Min =  -242  Ave =    -4.36  Std =    25.30
   Min TM value:-242 at an offset of 126.100 usec -> 2021-07-26T09:46:43.485.963.825 (XO: 17.485966377)
   Max TM value:240 at an offset of 123.950 usec -> 2021-07-26T09:46:43.485.961.675 (XO: 17.485964227)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 123.800 usec -> 2021-07-26T09:46:43.485.961.525 (XO: 17.485964077)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 125.913 usec -> 2021-07-26T09:46:43.485.963.637 (XO: 17.485966190)

 - Block number: 3 starting at 2021-07-26T09:47:04.455.092.725 (XO: 38.455094672 - 38.455400673, t=37.2 degC)  Rank: 00081
   TM values: Max =   235  Min =  -256  Ave =    -4.47  Std =    22.86
   Min TM value:-256 at an offset of 127.375 usec -> 2021-07-26T09:47:04.455.220.100 (XO: 38.455222048)
   Max TM value:235 at an offset of 127.063 usec -> 2021-07-26T09:47:04.455.219.787 (XO: 38.455221735)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 129.700 usec -> 2021-07-26T09:47:04.455.222.425 (XO: 38.455224373)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 127.175 usec -> 2021-07-26T09:47:04.455.219.900 (XO: 38.455221848)
