-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 26 18:25:15 2021 UTC
-- File 20210726_083238_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.663175 to 81.992143
-each separately from 3s to 81s and from 262011us to 992143us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.635171 in 1 packets: 18
      9.563597 in 1 packets: 36
      10.039809 in 1 packets: 54
      18.359103 in 1 packets: 72
      15.729554 in 1 packets: 90
MIU packet time from 2021-07-26T08:32:44.663.175.000 to 2021-07-26T08:34:02.992.143.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.635171 in 1 packets: 18
      9.563597 in 1 packets: 36
      10.039809 in 1 packets: 54
      18.359103 in 1 packets: 72
      15.729554 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9773774 to 82.9771148
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999158 in 1 packets: 18
      8.9999697 in 1 packets: 36
      9.9999664 in 1 packets: 54
      18.9999360 in 1 packets: 72
      14.9999495 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210726_083238_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.6405530 to 5.6408420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.6350870  -- OK for SL --
   MIU time from 2021-07-26T08:32:44.663.175.000 to 2021-07-26T08:32:44.663.464.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.635171

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.2759290 to 30.2762180
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5635670  -- OK for SL --
   MIU time from 2021-07-26T08:33:09.298.635.000 to 2021-07-26T08:33:09.298.924.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.563597

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.8397850 to 39.8400740
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0397750  -- OK for SL --
   MIU time from 2021-07-26T08:33:18.862.521.000 to 2021-07-26T08:33:18.862.810.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.039809

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.8798490 to 49.8801380
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3590390  -- OK for SL --
   MIU time from 2021-07-26T08:33:28.902.619.000 to 2021-07-26T08:33:28.902.908.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.359103

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.2391770 to 68.2394660
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7295030  -- OK for SL --
   MIU time from 2021-07-26T08:33:47.262.011.000 to 2021-07-26T08:33:47.262.300.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.729554

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 83.9689690 to 83.9692580
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-26T08:34:02.991.854.000 to 2021-07-26T08:34:02.992.143.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.640553000 corrected by 40bit 10MHz counter of 4.977377400 obtained at 1Hz counter of 3. Result: 3.663177855
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-26T08:32:44.663.034.725
   Max 1st sample MIU time among all SL packets: 2021-07-26T08:34:02.992.002.725
   Min 1st sample XO time among all SL packets: 3.663037579
   Max 1st sample XO time among all SL packets: 81.992006397
   TM values from -364 to 265 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-26T08:32:44.663.034.725 (XO: 3.663037579 - 3.663343580, t=37.2 degC)  Rank: 00071
   TM values: Max =   185  Min =  -229  Ave =    -4.46  Std =    20.80
   Min TM value:-229 at an offset of 105.988 usec -> 2021-07-26T08:32:44.663.140.712 (XO: 3.663143567)
   Max TM value:185 at an offset of 105.638 usec -> 2021-07-26T08:32:44.663.140.362 (XO: 3.663143217)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 108.300 usec -> 2021-07-26T08:32:44.663.143.025 (XO: 3.663145880)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 105.750 usec -> 2021-07-26T08:32:44.663.140.475 (XO: 3.663143330)

 - Block number: 2 starting at 2021-07-26T08:33:09.298.494.725 (XO: 28.298496510 - 28.298802511, t=37.2 degC)  Rank: 00076
   TM values: Max =   265  Min =  -364  Ave =    -4.55  Std =    38.24
   Min TM value:-364 at an offset of 100.425 usec -> 2021-07-26T08:33:09.298.595.150 (XO: 28.298596935)
   Max TM value:265 at an offset of 126.063 usec -> 2021-07-26T08:33:09.298.620.787 (XO: 28.298622573)
   Max dTM/dt [1/us]:2.88e+03 at an offset of 126.888 usec -> 2021-07-26T08:33:09.298.621.612 (XO: 28.298623398)
   Min dTM/dt [1/us]:-3.62e+03 at an offset of 126.113 usec -> 2021-07-26T08:33:09.298.620.837 (XO: 28.298622623)

 - Block number: 3 starting at 2021-07-26T08:33:18.862.380.725 (XO: 37.862384671 - 37.862690672, t=37.2 degC)  Rank: 00072
   TM values: Max =   161  Min =  -198  Ave =    -4.45  Std =    16.85
   Min TM value:-198 at an offset of 118.613 usec -> 2021-07-26T08:33:18.862.499.337 (XO: 37.862503284)
   Max TM value:161 at an offset of 121.225 usec -> 2021-07-26T08:33:18.862.501.950 (XO: 37.862505896)
   Max dTM/dt [1/us]:1.25e+03 at an offset of 121.050 usec -> 2021-07-26T08:33:18.862.501.775 (XO: 37.862505721)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 118.375 usec -> 2021-07-26T08:33:18.862.499.100 (XO: 37.862503046)

 - Block number: 4 starting at 2021-07-26T08:33:28.902.478.725 (XO: 47.902482493 - 47.902788494, t=37.2 degC)  Rank: 00089
   TM values: Max =   133  Min =  -309  Ave =    -4.38  Std =    22.38
   Min TM value:-309 at an offset of 92.000 usec -> 2021-07-26T08:33:28.902.570.725 (XO: 47.902574494)
   Max TM value:133 at an offset of 95.175 usec -> 2021-07-26T08:33:28.902.573.900 (XO: 47.902577669)
   Max dTM/dt [1/us]:7.12e+02 at an offset of 147.676 usec -> 2021-07-26T08:33:28.902.626.400 (XO: 47.902630169)
   Min dTM/dt [1/us]:-9.34e+02 at an offset of 91.875 usec -> 2021-07-26T08:33:28.902.570.600 (XO: 47.902574369)

 - Block number: 5 starting at 2021-07-26T08:33:47.261.870.725 (XO: 66.261872289 - 66.262178290, t=37.2 degC)  Rank: 00068
   TM values: Max =   165  Min =  -220  Ave =    -4.49  Std =    21.55
   Min TM value:-220 at an offset of 120.675 usec -> 2021-07-26T08:33:47.261.991.400 (XO: 66.261992965)
   Max TM value:165 at an offset of 123.200 usec -> 2021-07-26T08:33:47.261.993.925 (XO: 66.261995490)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 123.025 usec -> 2021-07-26T08:33:47.261.993.750 (XO: 66.261995315)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 120.438 usec -> 2021-07-26T08:33:47.261.991.162 (XO: 66.261992727)

 - Block number: 6 starting at 2021-07-26T08:34:02.991.713.725 (XO: 81.991717396 - 81.992023397, t=37.2 degC)  Rank: 00069
   TM values: Max =   168  Min =  -215  Ave =    -4.51  Std =    25.19
   Min TM value:-215 at an offset of 108.963 usec -> 2021-07-26T08:34:02.991.822.687 (XO: 81.991826359)
   Max TM value:168 at an offset of 108.675 usec -> 2021-07-26T08:34:02.991.822.400 (XO: 81.991826071)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 111.388 usec -> 2021-07-26T08:34:02.991.825.112 (XO: 81.991828784)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 108.788 usec -> 2021-07-26T08:34:02.991.822.512 (XO: 81.991826184)
