-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 24 00:26:08 2021 UTC
-- File 20210722_161421_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.981365 to 74.819334
-each separately from 7s to 74s and from 182179us to 981654us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.890708 in 1 packets: 18
      14.450689 in 1 packets: 36
      25.858550 in 1 packets: 54
      15.636577 in 1 packets: 72
MIU packet time from 2021-07-22T16:14:31.981.365.000 to 2021-07-22T16:15:38.819.334.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.890708 in 1 packets: 18
      14.450689 in 1 packets: 36
      25.858550 in 1 packets: 54
      15.636577 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8530396 to 75.8528151
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999631 in 1 packets: 18
      14.9999497 in 2 packets: 36, 72
      25.9999130 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 37 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      3 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 37 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      3 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210722_161421_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8344049 to 9.8346939
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.8906710  -- OK for SL --
   MIU time from 2021-07-22T16:14:31.981.365.000 to 2021-07-22T16:14:31.981.654.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.890708

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7253649 to 20.7256539
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4506390  -- OK for SL --
   MIU time from 2021-07-22T16:14:42.872.362.000 to 2021-07-22T16:14:42.872.651.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.450689

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.1762929 to 35.1765819
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8584630  -- OK for SL --
   MIU time from 2021-07-22T16:14:57.323.340.000 to 2021-07-22T16:14:57.323.629.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.858550

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.0350449 to 61.0353339
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6365270  -- OK for SL --
   MIU time from 2021-07-22T16:15:23.182.179.000 to 2021-07-22T16:15:23.182.468.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.636577

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 76.6718609 to 76.6721499
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-22T16:15:38.819.045.000 to 2021-07-22T16:15:38.819.334.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.834404900 corrected by 40bit 10MHz counter of 8.853039600 obtained at 1Hz counter of 7. Result: 7.981368539
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-22T16:14:31.981.224.725
   Max 1st sample MIU time among all SL packets: 2021-07-22T16:15:38.819.193.725
   Min 1st sample XO time among all SL packets: 7.981228263
   Max 1st sample XO time among all SL packets: 74.819197228
   TM values from -278 to 262 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-22T16:14:31.981.224.725 (XO: 7.981228263 - 7.981534264, t=37.2 degC)  Rank: 00080
   TM values: Max =   262  Min =  -237  Ave =    -4.47  Std =    20.58
   Min TM value:-237 at an offset of 124.088 usec -> 2021-07-22T16:14:31.981.348.812 (XO: 7.981352351)
   Max TM value:262 at an offset of 126.800 usec -> 2021-07-22T16:14:31.981.351.525 (XO: 7.981355063)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 126.663 usec -> 2021-07-22T16:14:31.981.351.387 (XO: 7.981354926)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 123.888 usec -> 2021-07-22T16:14:31.981.348.612 (XO: 7.981352151)

 - Block number: 2 starting at 2021-07-22T16:14:42.872.221.725 (XO: 18.872224803 - 18.872530804, t=37.2 degC)  Rank: 00087
   TM values: Max =   237  Min =  -258  Ave =    -4.45  Std =    22.01
   Min TM value:-258 at an offset of 121.725 usec -> 2021-07-22T16:14:42.872.343.450 (XO: 18.872346529)
   Max TM value:237 at an offset of 124.388 usec -> 2021-07-22T16:14:42.872.346.112 (XO: 18.872349191)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 124.213 usec -> 2021-07-22T16:14:42.872.345.937 (XO: 18.872349016)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 121.538 usec -> 2021-07-22T16:14:42.872.343.262 (XO: 18.872346341)

 - Block number: 3 starting at 2021-07-22T16:14:57.323.199.725 (XO: 33.323201292 - 33.323507293, t=37.2 degC)  Rank: 00080
   TM values: Max =   250  Min =  -255  Ave =    -4.28  Std =    25.64
   Min TM value:-255 at an offset of 177.813 usec -> 2021-07-22T16:14:57.323.377.538 (XO: 33.323379105)
   Max TM value:250 at an offset of 124.500 usec -> 2021-07-22T16:14:57.323.324.225 (XO: 33.323325792)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 175.363 usec -> 2021-07-22T16:14:57.323.375.088 (XO: 33.323376655)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 177.676 usec -> 2021-07-22T16:14:57.323.377.400 (XO: 33.323378967)

 - Block number: 4 starting at 2021-07-22T16:15:23.182.038.725 (XO: 59.182039844 - 59.182345845, t=37.2 degC)  Rank: 00079
   TM values: Max =   217  Min =  -269  Ave =    -4.32  Std =    25.69
   Min TM value:-269 at an offset of 180.363 usec -> 2021-07-22T16:15:23.182.219.088 (XO: 59.182220207)
   Max TM value:217 at an offset of 124.663 usec -> 2021-07-22T16:15:23.182.163.387 (XO: 59.182164507)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 182.763 usec -> 2021-07-22T16:15:23.182.221.488 (XO: 59.182222607)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 166.638 usec -> 2021-07-22T16:15:23.182.205.363 (XO: 59.182206482)

 - Block number: 5 starting at 2021-07-22T16:15:38.818.904.725 (XO: 74.818908227 - 74.819214228, t=37.2 degC)  Rank: 00084
   TM values: Max =   244  Min =  -278  Ave =    -4.44  Std =    25.14
   Min TM value:-278 at an offset of 111.038 usec -> 2021-07-22T16:15:38.819.015.762 (XO: 74.819019265)
   Max TM value:244 at an offset of 96.250 usec -> 2021-07-22T16:15:38.819.000.975 (XO: 74.819004478)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 96.138 usec -> 2021-07-22T16:15:38.819.000.862 (XO: 74.819004365)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 98.300 usec -> 2021-07-22T16:15:38.819.003.025 (XO: 74.819006528)
