-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 24 00:09:35 2021 UTC
-- File 20210722_160556_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.764098 to 20.574589
-each separately from 12s to 20s and from 574300us to 764387us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.809913 in 1 packets: 18
MIU packet time from 2021-07-22T16:06:11.764.098.000 to 2021-07-22T16:06:19.574.589.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.809913 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8662767 to 21.8662500
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999733 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210722_160556_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6303749 to 14.6306639
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.8098870  -- OK for SL --
   MIU time from 2021-07-22T16:06:11.764.098.000 to 2021-07-22T16:06:11.764.387.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.809913

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.4405509 to 22.4408399
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-22T16:06:19.574.300.000 to 2021-07-22T16:06:19.574.589.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.630374900 corrected by 40bit 10MHz counter of 13.866276700 obtained at 1Hz counter of 12. Result: 12.764100798
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-22T16:06:11.763.957.725
   Max 1st sample MIU time among all SL packets: 2021-07-22T16:06:19.574.448.725
   Min 1st sample XO time among all SL packets: 12.763960522
   Max 1st sample XO time among all SL packets: 20.574451521
   TM values from -289 to 250 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-22T16:06:11.763.957.725 (XO: 12.763960522 - 12.764266524, t=37.2 degC)  Rank: 00085
   TM values: Max =   218  Min =  -245  Ave =    -4.48  Std =    22.22
   Min TM value:-245 at an offset of 108.050 usec -> 2021-07-22T16:06:11.764.065.775 (XO: 12.764068573)
   Max TM value:218 at an offset of 105.925 usec -> 2021-07-22T16:06:11.764.063.650 (XO: 12.764066448)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 105.800 usec -> 2021-07-22T16:06:11.764.063.525 (XO: 12.764066323)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 107.913 usec -> 2021-07-22T16:06:11.764.065.637 (XO: 12.764068435)

 - Block number: 2 starting at 2021-07-22T16:06:19.574.159.725 (XO: 20.574162520 - 20.574468521, t=37.2 degC)  Rank: 00080
   TM values: Max =   250  Min =  -289  Ave =    -4.41  Std =    32.04
   Min TM value:-289 at an offset of 189.813 usec -> 2021-07-22T16:06:19.574.349.538 (XO: 20.574352333)
   Max TM value:250 at an offset of 119.913 usec -> 2021-07-22T16:06:19.574.279.637 (XO: 20.574282433)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 122.563 usec -> 2021-07-22T16:06:19.574.282.287 (XO: 20.574285083)
   Min dTM/dt [1/us]:-2.82e+03 at an offset of 120.038 usec -> 2021-07-22T16:06:19.574.279.762 (XO: 20.574282558)
