-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 23 18:56:41 2021 UTC
-- File 20210722_124758_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.652381 to 22.762870
-each separately from 10s to 22s and from 652381us to 762870us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.109911 in 1 packets: 18
MIU packet time from 2021-07-22T12:48:11.652.381.000 to 2021-07-22T12:48:23.762.870.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.109911 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9303120 to 23.9302722
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999602 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210722_124758_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.5826941 to 12.5829831
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1098710  -- OK for SL --
   MIU time from 2021-07-22T12:48:11.652.381.000 to 2021-07-22T12:48:11.652.670.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.109911

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6928541 to 24.6931431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-22T12:48:23.762.581.000 to 2021-07-22T12:48:23.762.870.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.582694100 corrected by 40bit 10MHz counter of 11.930312000 obtained at 1Hz counter of 10. Result: 10.652384253
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-22T12:48:11.652.240.725
   Max 1st sample MIU time among all SL packets: 2021-07-22T12:48:23.762.729.725
   Min 1st sample XO time among all SL packets: 10.652243977
   Max 1st sample XO time among all SL packets: 22.762733142
   TM values from -275 to 258 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-22T12:48:11.652.240.725 (XO: 10.652243977 - 10.652549978, t=37.2 degC)  Rank: 00080
   TM values: Max =   244  Min =  -227  Ave =    -4.36  Std =    29.12
   Min TM value:-227 at an offset of 120.863 usec -> 2021-07-22T12:48:11.652.361.587 (XO: 10.652364840)
   Max TM value:244 at an offset of 123.625 usec -> 2021-07-22T12:48:11.652.364.350 (XO: 10.652367603)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 234.038 usec -> 2021-07-22T12:48:11.652.474.763 (XO: 10.652478016)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 120.713 usec -> 2021-07-22T12:48:11.652.361.437 (XO: 10.652364690)

 - Block number: 2 starting at 2021-07-22T12:48:23.762.440.725 (XO: 22.762444141 - 22.762750142, t=37.2 degC)  Rank: 00083
   TM values: Max =   258  Min =  -275  Ave =    -4.25  Std =    25.73
   Min TM value:-275 at an offset of 129.513 usec -> 2021-07-22T12:48:23.762.570.237 (XO: 22.762573654)
   Max TM value:258 at an offset of 88.675 usec -> 2021-07-22T12:48:23.762.529.400 (XO: 22.762532816)
   Max dTM/dt [1/us]:2.55e+03 at an offset of 91.325 usec -> 2021-07-22T12:48:23.762.532.050 (XO: 22.762535466)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 88.788 usec -> 2021-07-22T12:48:23.762.529.512 (XO: 22.762532929)
