-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 23 18:52:59 2021 UTC
-- File 20210722_124621_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.612780 to 23.142691
-each separately from 2s to 23s and from 142402us to 613069us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.529333 in 1 packets: 18
MIU packet time from 2021-07-22T12:46:26.612.780.000 to 2021-07-22T12:46:47.142.691.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.529333 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7342535 to 24.7341837
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999302 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210722_124621_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3470345 to 4.3473235
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5292630  -- OK for SL --
   MIU time from 2021-07-22T12:46:26.612.780.000 to 2021-07-22T12:46:26.613.069.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.529333

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.8765865 to 24.8768755
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-22T12:46:47.142.402.000 to 2021-07-22T12:46:47.142.691.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.347034500 corrected by 40bit 10MHz counter of 3.734253500 obtained at 1Hz counter of 2. Result: 2.612783022
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-07-22T12:46:26.612.639.725
   Max 1st sample MIU time among all SL packets: 2021-07-22T12:46:47.142.550.725
   Min 1st sample XO time among all SL packets: 2.612642747
   Max 1st sample XO time among all SL packets: 23.142551981
   TM values from -283 to 246 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-22T12:46:26.612.639.725 (XO: 2.612642747 - 2.612948748, t=37.2 degC)  Rank: 00083
   TM values: Max =   246  Min =  -244  Ave =    -4.38  Std =    19.37
   Min TM value:-244 at an offset of 122.188 usec -> 2021-07-22T12:46:26.612.761.912 (XO: 2.612764935)
   Max TM value:246 at an offset of 124.888 usec -> 2021-07-22T12:46:26.612.764.612 (XO: 2.612767635)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 124.775 usec -> 2021-07-22T12:46:26.612.764.500 (XO: 2.612767522)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 110.238 usec -> 2021-07-22T12:46:26.612.749.962 (XO: 2.612752985)

 - Block number: 2 starting at 2021-07-22T12:46:47.142.261.725 (XO: 23.142262980 - 23.142568981, t=37.2 degC)  Rank: 00080
   TM values: Max =   212  Min =  -283  Ave =    -4.36  Std =    19.77
   Min TM value:-283 at an offset of 121.713 usec -> 2021-07-22T12:46:47.142.383.437 (XO: 23.142384693)
   Max TM value:212 at an offset of 119.413 usec -> 2021-07-22T12:46:47.142.381.137 (XO: 23.142382393)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 119.250 usec -> 2021-07-22T12:46:47.142.380.975 (XO: 23.142382231)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 121.525 usec -> 2021-07-22T12:46:47.142.383.250 (XO: 23.142384506)
