-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 18:40:22 2021 UTC
-- File 20210716_085133_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.424069 to 44.773248
-each separately from 8s to 44s and from 424069us to 833995us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.409348 in 1 packets: 18
      24.938964 in 1 packets: 36
MIU packet time from 2021-07-16T08:51:44.424.069.000 to 2021-07-16T08:52:20.773.248.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.409348 in 1 packets: 18
      24.938964 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8207820 to 45.8206607
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999632 in 1 packets: 18
      24.9999155 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210716_085133_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2448520 to 10.2451410
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.4093110  -- OK for SL --
   MIU time from 2021-07-16T08:51:44.424.069.000 to 2021-07-16T08:51:44.424.358.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.409348

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6544520 to 21.6547410
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9388790  -- OK for SL --
   MIU time from 2021-07-16T08:51:55.833.706.000 to 2021-07-16T08:51:55.833.995.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.938964

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.5936200 to 46.5939090
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-16T08:52:20.772.959.000 to 2021-07-16T08:52:20.773.248.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.244852000 corrected by 40bit 10MHz counter of 9.820782000 obtained at 1Hz counter of 8. Result: 8.424071442
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-16T08:51:44.423.928.725
   Max 1st sample MIU time among all SL packets: 2021-07-16T08:52:20.773.107.725
   Min 1st sample XO time among all SL packets: 8.423931166
   Max 1st sample XO time among all SL packets: 44.773110654
   TM values from -237 to 222 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-16T08:51:44.423.928.725 (XO: 8.423931166 - 8.424237167, t=37.2 degC)  Rank: 00080
   TM values: Max =   222  Min =  -237  Ave =    -4.50  Std =    22.19
   Min TM value:-237 at an offset of 122.413 usec -> 2021-07-16T08:51:44.424.051.137 (XO: 8.424053579)
   Max TM value:222 at an offset of 122.113 usec -> 2021-07-16T08:51:44.424.050.837 (XO: 8.424053279)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 124.750 usec -> 2021-07-16T08:51:44.424.053.475 (XO: 8.424055917)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 122.213 usec -> 2021-07-16T08:51:44.424.050.937 (XO: 8.424053379)

 - Block number: 2 starting at 2021-07-16T08:51:55.833.565.725 (XO: 19.833569359 - 19.833875360, t=37.2 degC)  Rank: 00081
   TM values: Max =   184  Min =  -234  Ave =    -4.52  Std =    20.25
   Min TM value:-234 at an offset of 112.813 usec -> 2021-07-16T08:51:55.833.678.537 (XO: 19.833682172)
   Max TM value:184 at an offset of 123.488 usec -> 2021-07-16T08:51:55.833.689.212 (XO: 19.833692847)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 123.313 usec -> 2021-07-16T08:51:55.833.689.037 (XO: 19.833692672)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 125.438 usec -> 2021-07-16T08:51:55.833.691.162 (XO: 19.833694797)

 - Block number: 3 starting at 2021-07-16T08:52:20.772.818.725 (XO: 44.772821653 - 44.773127654, t=37.2 degC)  Rank: 00082
   TM values: Max =   166  Min =  -220  Ave =    -4.46  Std =    22.86
   Min TM value:-220 at an offset of 123.675 usec -> 2021-07-16T08:52:20.772.942.400 (XO: 44.772945328)
   Max TM value:166 at an offset of 123.313 usec -> 2021-07-16T08:52:20.772.942.037 (XO: 44.772944966)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 126.075 usec -> 2021-07-16T08:52:20.772.944.800 (XO: 44.772947728)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 123.488 usec -> 2021-07-16T08:52:20.772.942.212 (XO: 44.772945141)
