-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 12:53:56 2021 UTC
-- File 20210715_140410_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.607915 to 30.219699
-each separately from 4s to 30s and from 198583us to 608204us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.590379 in 1 packets: 18
      8.020538 in 1 packets: 36
MIU packet time from 2021-07-15T14:04:17.607.915.000 to 2021-07-15T14:04:43.219.699.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.590379 in 1 packets: 18
      8.020538 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8549663 to 31.8548793
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999398 in 1 packets: 18
      7.9999732 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_140410_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4628823 to 6.4631713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5903190  -- OK for SL --
   MIU time from 2021-07-15T14:04:17.607.915.000 to 2021-07-15T14:04:17.608.204.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.590379

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.0534903 to 24.0537793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0205110  -- OK for SL --
   MIU time from 2021-07-15T14:04:35.198.583.000 to 2021-07-15T14:04:35.198.872.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.020538

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0742903 to 32.0745793
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T14:04:43.219.410.000 to 2021-07-15T14:04:43.219.699.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.462882300 corrected by 40bit 10MHz counter of 5.854966300 obtained at 1Hz counter of 4. Result: 4.607918006
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T14:04:17.607.774.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T14:04:43.219.558.725
   Min 1st sample XO time among all SL packets: 4.607777731
   Max 1st sample XO time among all SL packets: 30.219560472
   TM values from -283 to 265 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T14:04:17.607.774.725 (XO: 4.607777731 - 4.608083732, t=37.2 degC)  Rank: 00081
   TM values: Max =   248  Min =  -282  Ave =    -4.58  Std =    21.94
   Min TM value:-282 at an offset of 113.825 usec -> 2021-07-15T14:04:17.607.888.550 (XO: 4.607891556)
   Max TM value:248 at an offset of 113.588 usec -> 2021-07-15T14:04:17.607.888.312 (XO: 4.607891319)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 111.625 usec -> 2021-07-15T14:04:17.607.886.350 (XO: 4.607889356)
   Min dTM/dt [1/us]:-2.57e+03 at an offset of 113.700 usec -> 2021-07-15T14:04:17.607.888.425 (XO: 4.607891431)

 - Block number: 2 starting at 2021-07-15T14:04:35.198.442.725 (XO: 22.198444600 - 22.198750601, t=37.2 degC)  Rank: 00091
   TM values: Max =   265  Min =  -283  Ave =    -4.52  Std =    32.26
   Min TM value:-283 at an offset of 114.800 usec -> 2021-07-15T14:04:35.198.557.525 (XO: 22.198559400)
   Max TM value:265 at an offset of 114.563 usec -> 2021-07-15T14:04:35.198.557.287 (XO: 22.198559163)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 117.200 usec -> 2021-07-15T14:04:35.198.559.925 (XO: 22.198561800)
   Min dTM/dt [1/us]:-2.88e+03 at an offset of 114.675 usec -> 2021-07-15T14:04:35.198.557.400 (XO: 22.198559275)

 - Block number: 3 starting at 2021-07-15T14:04:43.219.269.725 (XO: 30.219271471 - 30.219577472, t=37.2 degC)  Rank: 00079
   TM values: Max =   233  Min =  -251  Ave =    -4.60  Std =    22.25
   Min TM value:-251 at an offset of 126.963 usec -> 2021-07-15T14:04:43.219.396.687 (XO: 30.219398433)
   Max TM value:233 at an offset of 129.588 usec -> 2021-07-15T14:04:43.219.399.312 (XO: 30.219401058)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 129.413 usec -> 2021-07-15T14:04:43.219.399.137 (XO: 30.219400883)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 126.788 usec -> 2021-07-15T14:04:43.219.396.512 (XO: 30.219398258)
