-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 12:29:43 2021 UTC
-- File 20210715_133757_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.530910 to 39.939253
-each separately from 9s to 39s and from 519848us to 939253us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.988649 in 1 packets: 18
      13.418827 in 1 packets: 36
MIU packet time from 2021-07-15T13:38:09.530.910.000 to 2021-07-15T13:38:39.939.253.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.988649 in 1 packets: 18
      13.418827 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8729695 to 40.8728685
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999428 in 1 packets: 18
      12.9999562 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_133757_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.4038806 to 11.4041696
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9885910  -- OK for SL --
   MIU time from 2021-07-15T13:38:09.530.910.000 to 2021-07-15T13:38:09.531.199.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.988649

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3927606 to 28.3930496
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4187830  -- OK for SL --
   MIU time from 2021-07-15T13:38:26.519.848.000 to 2021-07-15T13:38:26.520.137.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.418827

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.8118326 to 41.8121216
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T13:38:39.938.964.000 to 2021-07-15T13:38:39.939.253.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.403880600 corrected by 40bit 10MHz counter of 10.872969500 obtained at 1Hz counter of 9. Result: 9.530912905
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-15T13:38:09.530.769.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T13:38:39.939.112.725
   Min 1st sample XO time among all SL packets: 9.530772630
   Max 1st sample XO time among all SL packets: 39.939116018
   TM values from -301 to 258 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T13:38:09.530.769.725 (XO: 9.530772630 - 9.531078631, t=37.2 degC)  Rank: 00079
   TM values: Max =   258  Min =  -301  Ave =    -4.43  Std =    28.18
   Min TM value:-301 at an offset of 129.663 usec -> 2021-07-15T13:38:09.530.899.387 (XO: 9.530902293)
   Max TM value:258 at an offset of 132.313 usec -> 2021-07-15T13:38:09.530.902.037 (XO: 9.530904943)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 132.200 usec -> 2021-07-15T13:38:09.530.901.925 (XO: 9.530904830)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 129.550 usec -> 2021-07-15T13:38:09.530.899.275 (XO: 9.530902180)

 - Block number: 2 starting at 2021-07-15T13:38:26.519.707.725 (XO: 26.519709792 - 26.520015793, t=37.2 degC)  Rank: 00090
   TM values: Max =   219  Min =  -288  Ave =    -4.49  Std =    24.63
   Min TM value:-288 at an offset of 126.438 usec -> 2021-07-15T13:38:26.519.834.162 (XO: 26.519836230)
   Max TM value:219 at an offset of 129.038 usec -> 2021-07-15T13:38:26.519.836.762 (XO: 26.519838830)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 128.788 usec -> 2021-07-15T13:38:26.519.836.512 (XO: 26.519838580)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 126.250 usec -> 2021-07-15T13:38:26.519.833.975 (XO: 26.519836042)

 - Block number: 3 starting at 2021-07-15T13:38:39.938.823.725 (XO: 39.938827017 - 39.939133018, t=37.2 degC)  Rank: 00080
   TM values: Max =   253  Min =  -271  Ave =    -4.46  Std =    22.98
   Min TM value:-271 at an offset of 90.788 usec -> 2021-07-15T13:38:39.938.914.512 (XO: 39.938917805)
   Max TM value:253 at an offset of 90.488 usec -> 2021-07-15T13:38:39.938.914.212 (XO: 39.938917505)
   Max dTM/dt [1/us]:2.81e+03 at an offset of 93.125 usec -> 2021-07-15T13:38:39.938.916.850 (XO: 39.938920142)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 90.600 usec -> 2021-07-15T13:38:39.938.914.325 (XO: 39.938917617)
