-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 12:12:23 2021 UTC
-- File 20210715_131359_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.249059 to 29.388093
-each separately from 12s to 29s and from 249059us to 388093us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.138456 in 1 packets: 18
MIU packet time from 2021-07-15T13:14:14.249.059.000 to 2021-07-15T13:14:31.388.093.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.138456 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8229451 to 30.8228877
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999426 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_131359_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0720050 to 14.0722940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1383990  -- OK for SL --
   MIU time from 2021-07-15T13:14:14.249.059.000 to 2021-07-15T13:14:14.249.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.138456

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.2106930 to 31.2109820
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T13:14:31.387.804.000 to 2021-07-15T13:14:31.388.093.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.072005000 corrected by 40bit 10MHz counter of 13.822945100 obtained at 1Hz counter of 12. Result: 12.249060722
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T13:14:14.248.918.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T13:14:31.387.952.725
   Min 1st sample XO time among all SL packets: 12.248920446
   Max 1st sample XO time among all SL packets: 29.387955344
   TM values from -261 to 226 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T13:14:14.248.918.725 (XO: 12.248920446 - 12.249226447, t=37.2 degC)  Rank: 00082
   TM values: Max =   226  Min =  -261  Ave =    -4.57  Std =    27.36
   Min TM value:-261 at an offset of 109.063 usec -> 2021-07-15T13:14:14.249.027.787 (XO: 12.249029509)
   Max TM value:226 at an offset of 108.763 usec -> 2021-07-15T13:14:14.249.027.487 (XO: 12.249029209)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 111.425 usec -> 2021-07-15T13:14:14.249.030.150 (XO: 12.249031872)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 108.875 usec -> 2021-07-15T13:14:14.249.027.600 (XO: 12.249029322)

 - Block number: 2 starting at 2021-07-15T13:14:31.387.663.725 (XO: 29.387666343 - 29.387972344, t=37.2 degC)  Rank: 00079
   TM values: Max =   215  Min =  -235  Ave =    -4.56  Std =    19.65
   Min TM value:-235 at an offset of 105.913 usec -> 2021-07-15T13:14:31.387.769.637 (XO: 29.387772256)
   Max TM value:215 at an offset of 108.525 usec -> 2021-07-15T13:14:31.387.772.250 (XO: 29.387774868)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 108.413 usec -> 2021-07-15T13:14:31.387.772.137 (XO: 29.387774756)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 105.688 usec -> 2021-07-15T13:14:31.387.769.412 (XO: 29.387772031)
