-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 11:31:06 2021 UTC
-- File 20210715_122156_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.613369 to 73.970925
-each separately from 5s to 73s and from 521837us to 982889us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.368942 in 1 packets: 18
      15.538948 in 1 packets: 36
      13.209354 in 1 packets: 54
      25.238867 in 1 packets: 72
MIU packet time from 2021-07-15T12:22:04.613.369.000 to 2021-07-15T12:23:12.970.925.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.368942 in 1 packets: 18
      15.538948 in 1 packets: 36
      13.209354 in 1 packets: 54
      25.238867 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9194831 to 74.9192564
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
      15.9999467 in 1 packets: 36
      12.9999567 in 1 packets: 54
      24.9999166 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_122156_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.5328525 to 7.5331415
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3688950  -- OK for SL --
   MIU time from 2021-07-15T12:22:04.613.369.000 to 2021-07-15T12:22:04.613.658.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.368942

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9020365 to 21.9023255
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5388950  -- OK for SL --
   MIU time from 2021-07-15T12:22:18.982.600.000 to 2021-07-15T12:22:18.982.889.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.538948

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4412205 to 37.4415095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2093110  -- OK for SL --
   MIU time from 2021-07-15T12:22:34.521.837.000 to 2021-07-15T12:22:34.522.126.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.209354

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.6508205 to 50.6511095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.2387830  -- OK for SL --
   MIU time from 2021-07-15T12:22:47.731.480.000 to 2021-07-15T12:22:47.731.769.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.238867

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.8898925 to 75.8901815
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T12:23:12.970.636.000 to 2021-07-15T12:23:12.970.925.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.532852500 corrected by 40bit 10MHz counter of 6.919483100 obtained at 1Hz counter of 5. Result: 5.613371424
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T12:22:04.613.228.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T12:23:12.970.784.725
   Min 1st sample XO time among all SL packets: 5.613231149
   Max 1st sample XO time among all SL packets: 73.970788029
   TM values from -277 to 239 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T12:22:04.613.228.725 (XO: 5.613231149 - 5.613537150, t=37.2 degC)  Rank: 00080
   TM values: Max =   194  Min =  -277  Ave =    -4.43  Std =    20.97
   Min TM value:-277 at an offset of 122.513 usec -> 2021-07-15T12:22:04.613.351.237 (XO: 5.613353662)
   Max TM value:194 at an offset of 122.213 usec -> 2021-07-15T12:22:04.613.350.937 (XO: 5.613353362)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 124.925 usec -> 2021-07-15T12:22:04.613.353.650 (XO: 5.613356074)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 122.413 usec -> 2021-07-15T12:22:04.613.351.137 (XO: 5.613353562)

 - Block number: 2 starting at 2021-07-15T12:22:18.982.459.725 (XO: 19.982463067 - 19.982769068, t=37.2 degC)  Rank: 00084
   TM values: Max =   239  Min =  -266  Ave =    -4.51  Std =    21.16
   Min TM value:-266 at an offset of 114.000 usec -> 2021-07-15T12:22:18.982.573.725 (XO: 19.982577068)
   Max TM value:239 at an offset of 116.675 usec -> 2021-07-15T12:22:18.982.576.400 (XO: 19.982579743)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 111.813 usec -> 2021-07-15T12:22:18.982.571.537 (XO: 19.982574880)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 113.875 usec -> 2021-07-15T12:22:18.982.573.600 (XO: 19.982576943)

 - Block number: 3 starting at 2021-07-15T12:22:34.521.696.725 (XO: 35.521698899 - 35.522004900, t=37.2 degC)  Rank: 00080
   TM values: Max =   212  Min =  -264  Ave =    -4.54  Std =    18.00
   Min TM value:-264 at an offset of 115.538 usec -> 2021-07-15T12:22:34.521.812.262 (XO: 35.521814437)
   Max TM value:212 at an offset of 115.263 usec -> 2021-07-15T12:22:34.521.811.987 (XO: 35.521814162)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 113.075 usec -> 2021-07-15T12:22:34.521.809.800 (XO: 35.521811974)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 115.375 usec -> 2021-07-15T12:22:34.521.812.100 (XO: 35.521814274)

 - Block number: 4 starting at 2021-07-15T12:22:47.731.339.725 (XO: 48.731342912 - 48.731648913, t=37.2 degC)  Rank: 00079
   TM values: Max =   220  Min =  -243  Ave =    -4.53  Std =    19.59
   Min TM value:-243 at an offset of 125.900 usec -> 2021-07-15T12:22:47.731.465.625 (XO: 48.731468812)
   Max TM value:220 at an offset of 128.550 usec -> 2021-07-15T12:22:47.731.468.275 (XO: 48.731471462)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 128.388 usec -> 2021-07-15T12:22:47.731.468.112 (XO: 48.731471300)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 125.675 usec -> 2021-07-15T12:22:47.731.465.400 (XO: 48.731468587)

 - Block number: 5 starting at 2021-07-15T12:23:12.970.495.725 (XO: 73.970499028 - 73.970805029, t=37.2 degC)  Rank: 00079
   TM values: Max =   226  Min =  -277  Ave =    -4.51  Std =    27.63
   Min TM value:-277 at an offset of 155.226 usec -> 2021-07-15T12:23:12.970.650.950 (XO: 73.970654253)
   Max TM value:226 at an offset of 126.413 usec -> 2021-07-15T12:23:12.970.622.137 (XO: 73.970625441)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 129.088 usec -> 2021-07-15T12:23:12.970.624.812 (XO: 73.970628116)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 126.525 usec -> 2021-07-15T12:23:12.970.622.250 (XO: 73.970625553)
