-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 11:05:21 2021 UTC
-- File 20210715_121030_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.490141 to 21.320893
-each separately from 2s to 21s and from 320604us to 490430us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.830174 in 1 packets: 18
MIU packet time from 2021-07-15T12:10:35.490.141.000 to 2021-07-15T12:10:54.320.893.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.830174 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8908182 to 22.8907550
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999368 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_121030_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3809604 to 4.3812494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8301110  -- OK for SL --
   MIU time from 2021-07-15T12:10:35.490.141.000 to 2021-07-15T12:10:35.490.430.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.830174

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2113604 to 23.2116494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T12:10:54.320.604.000 to 2021-07-15T12:10:54.320.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.380960400 corrected by 40bit 10MHz counter of 3.890818200 obtained at 1Hz counter of 2. Result: 2.490143866
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T12:10:35.490.000.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T12:10:54.320.752.725
   Min 1st sample XO time among all SL packets: 2.490003591
   Max 1st sample XO time among all SL packets: 21.320755183
   TM values from -258 to 226 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T12:10:35.490.000.725 (XO: 2.490003591 - 2.490309592, t=37.2 degC)  Rank: 00078
   TM values: Max =   226  Min =  -258  Ave =    -4.51  Std =    22.86
   Min TM value:-258 at an offset of 123.813 usec -> 2021-07-15T12:10:35.490.124.537 (XO: 2.490127404)
   Max TM value:226 at an offset of 123.525 usec -> 2021-07-15T12:10:35.490.124.250 (XO: 2.490127116)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 121.388 usec -> 2021-07-15T12:10:35.490.122.112 (XO: 2.490124979)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 123.650 usec -> 2021-07-15T12:10:35.490.124.375 (XO: 2.490127241)

 - Block number: 2 starting at 2021-07-15T12:10:54.320.463.725 (XO: 21.320466183 - 21.320772184, t=37.2 degC)  Rank: 00082
   TM values: Max =   215  Min =  -231  Ave =    -4.54  Std =    20.21
   Min TM value:-231 at an offset of 118.813 usec -> 2021-07-15T12:10:54.320.582.537 (XO: 21.320584995)
   Max TM value:215 at an offset of 121.500 usec -> 2021-07-15T12:10:54.320.585.225 (XO: 21.320587683)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 121.350 usec -> 2021-07-15T12:10:54.320.585.075 (XO: 21.320587533)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 118.613 usec -> 2021-07-15T12:10:54.320.582.337 (XO: 21.320584795)
