-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 10:27:57 2021 UTC
-- File 20210715_115419_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.735365 to 22.775943
-each separately from 12s to 22s and from 735365us to 775943us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.040000 in 1 packets: 18
MIU packet time from 2021-07-15T11:54:34.735.365.000 to 2021-07-15T11:54:44.775.943.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.040000 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8854506 to 22.8854171
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999665 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_115419_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.6208162 to 13.6211052
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0399670  -- OK for SL --
   MIU time from 2021-07-15T11:54:34.735.365.000 to 2021-07-15T11:54:34.735.654.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.040000

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6610722 to 23.6613612
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T11:54:44.775.654.000 to 2021-07-15T11:54:44.775.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 30 XO calibration points at 1Hz counter from 0 to 29
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.620816200 corrected by 40bit 10MHz counter of 12.885450600 obtained at 1Hz counter of 12. Result: 12.735368027
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T11:54:34.735.224.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T11:54:44.775.802.725
   Min 1st sample XO time among all SL packets: 12.735227751
   Max 1st sample XO time among all SL packets: 22.775806463
   TM values from -276 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T11:54:34.735.224.725 (XO: 12.735227751 - 12.735533752, t=37.2 degC)  Rank: 00084
   TM values: Max =   201  Min =  -244  Ave =    -4.44  Std =    21.32
   Min TM value:-244 at an offset of 118.675 usec -> 2021-07-15T11:54:34.735.343.400 (XO: 12.735346427)
   Max TM value:201 at an offset of 121.263 usec -> 2021-07-15T11:54:34.735.345.987 (XO: 12.735349014)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 121.063 usec -> 2021-07-15T11:54:34.735.345.787 (XO: 12.735348814)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 118.475 usec -> 2021-07-15T11:54:34.735.343.200 (XO: 12.735346227)

 - Block number: 2 starting at 2021-07-15T11:54:44.775.513.725 (XO: 22.775517462 - 22.775823463, t=37.2 degC)  Rank: 00081
   TM values: Max =   228  Min =  -276  Ave =    -4.47  Std =    23.93
   Min TM value:-276 at an offset of 107.125 usec -> 2021-07-15T11:54:44.775.620.850 (XO: 22.775624587)
   Max TM value:228 at an offset of 106.813 usec -> 2021-07-15T11:54:44.775.620.537 (XO: 22.775624275)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 109.613 usec -> 2021-07-15T11:54:44.775.623.337 (XO: 22.775627075)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 106.963 usec -> 2021-07-15T11:54:44.775.620.687 (XO: 22.775624425)
