-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 09:11:37 2021 UTC
-- File 20210715_104913_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.681557 to 63.759763
-each separately from 11s to 63s and from 11840us to 949960us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329994 in 1 packets: 18
      9.108989 in 1 packets: 36
      17.828264 in 1 packets: 54
      12.809514 in 1 packets: 72
MIU packet time from 2021-07-15T10:49:27.681.557.000 to 2021-07-15T10:50:19.759.763.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.329994 in 1 packets: 18
      9.108989 in 1 packets: 36
      17.828264 in 1 packets: 54
      12.809514 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8373856 to 64.8372117
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999566 in 1 packets: 18
      8.9999699 in 1 packets: 36
      16.9999432 in 1 packets: 54
      12.9999564 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_104913_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.5189428 to 13.5192318
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3299510  -- OK for SL --
   MIU time from 2021-07-15T10:49:27.681.557.000 to 2021-07-15T10:49:27.681.846.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.329994

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.8491828 to 25.8494718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1089590  -- OK for SL --
   MIU time from 2021-07-15T10:49:40.011.840.000 to 2021-07-15T10:49:40.012.129.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.108989

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9584308 to 34.9587198
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8282070  -- OK for SL --
   MIU time from 2021-07-15T10:49:49.121.118.000 to 2021-07-15T10:49:49.121.407.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.828264

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.7869268 to 52.7872158
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8094710  -- OK for SL --
   MIU time from 2021-07-15T10:50:06.949.671.000 to 2021-07-15T10:50:06.949.960.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.809514

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.5966868 to 65.5969758
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T10:50:19.759.474.000 to 2021-07-15T10:50:19.759.763.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.518942800 corrected by 40bit 10MHz counter of 12.837385600 obtained at 1Hz counter of 11. Result: 11.681559449
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T10:49:27.681.416.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T10:50:19.759.622.725
   Min 1st sample XO time among all SL packets: 11.681419174
   Max 1st sample XO time among all SL packets: 63.759626408
   TM values from -269 to 245 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T10:49:27.681.416.725 (XO: 11.681419174 - 11.681725175, t=37.2 degC)  Rank: 00078
   TM values: Max =   235  Min =  -256  Ave =    -4.52  Std =    19.95
   Min TM value:-256 at an offset of 90.538 usec -> 2021-07-15T10:49:27.681.507.262 (XO: 11.681509711)
   Max TM value:235 at an offset of 90.250 usec -> 2021-07-15T10:49:27.681.506.975 (XO: 11.681509424)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 92.925 usec -> 2021-07-15T10:49:27.681.509.650 (XO: 11.681512099)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 90.375 usec -> 2021-07-15T10:49:27.681.507.100 (XO: 11.681509549)

 - Block number: 2 starting at 2021-07-15T10:49:40.011.699.725 (XO: 24.011700365 - 24.012006366, t=37.2 degC)  Rank: 00083
   TM values: Max =   226  Min =  -238  Ave =    -4.39  Std =    20.14
   Min TM value:-238 at an offset of 111.725 usec -> 2021-07-15T10:49:40.011.811.450 (XO: 24.011812090)
   Max TM value:226 at an offset of 114.375 usec -> 2021-07-15T10:49:40.011.814.100 (XO: 24.011814740)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 114.263 usec -> 2021-07-15T10:49:40.011.813.987 (XO: 24.011814628)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 111.563 usec -> 2021-07-15T10:49:40.011.811.287 (XO: 24.011811928)

 - Block number: 3 starting at 2021-07-15T10:49:49.120.977.725 (XO: 33.120978824 - 33.121284825, t=37.2 degC)  Rank: 00080
   TM values: Max =   245  Min =  -269  Ave =    -4.51  Std =    28.70
   Min TM value:-269 at an offset of 113.125 usec -> 2021-07-15T10:49:49.121.090.850 (XO: 33.121091950)
   Max TM value:245 at an offset of 112.875 usec -> 2021-07-15T10:49:49.121.090.600 (XO: 33.121091700)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 115.538 usec -> 2021-07-15T10:49:49.121.093.262 (XO: 33.121094362)
   Min dTM/dt [1/us]:-2.63e+03 at an offset of 112.988 usec -> 2021-07-15T10:49:49.121.090.712 (XO: 33.121091812)

 - Block number: 4 starting at 2021-07-15T10:50:06.949.530.725 (XO: 50.949534453 - 50.949840454, t=37.2 degC)  Rank: 00081
   TM values: Max =   229  Min =  -253  Ave =    -4.48  Std =    20.16
   Min TM value:-253 at an offset of 126.013 usec -> 2021-07-15T10:50:06.949.656.737 (XO: 50.949660466)
   Max TM value:229 at an offset of 128.700 usec -> 2021-07-15T10:50:06.949.659.425 (XO: 50.949663154)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 128.588 usec -> 2021-07-15T10:50:06.949.659.312 (XO: 50.949663041)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 125.888 usec -> 2021-07-15T10:50:06.949.656.612 (XO: 50.949660341)

 - Block number: 5 starting at 2021-07-15T10:50:19.759.333.725 (XO: 63.759337407 - 63.759643408, t=37.2 degC)  Rank: 00079
   TM values: Max =   223  Min =  -243  Ave =    -4.47  Std =    22.31
   Min TM value:-243 at an offset of 127.638 usec -> 2021-07-15T10:50:19.759.461.362 (XO: 63.759465045)
   Max TM value:223 at an offset of 125.075 usec -> 2021-07-15T10:50:19.759.458.800 (XO: 63.759462482)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 127.738 usec -> 2021-07-15T10:50:19.759.461.462 (XO: 63.759465145)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 125.175 usec -> 2021-07-15T10:50:19.759.458.900 (XO: 63.759462582)
