-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov 15 04:44:59 2021 UTC
-- File 20210715_030904_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.315708 to 25.494605
-each separately from 6s to 25s and from 315708us to 494605us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.178319 in 1 packets: 18
MIU packet time from 2021-07-15T03:09:13.315.708.000 to 2021-07-15T03:09:32.494.605.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.178319 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9474557 to 26.9473919
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999362 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210715_030904_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2631651 to 8.2634541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1782550  -- OK for SL --
   MIU time from 2021-07-15T03:09:13.315.708.000 to 2021-07-15T03:09:13.315.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.178319

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4417091 to 27.4419981
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-15T03:09:32.494.316.000 to 2021-07-15T03:09:32.494.605.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.263165100 corrected by 40bit 10MHz counter of 7.947455700 obtained at 1Hz counter of 6. Result: 6.315710442
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-15T03:09:13.315.567.725
   Max 1st sample MIU time among all SL packets: 2021-07-15T03:09:32.494.464.725
   Min 1st sample XO time among all SL packets: 6.315570166
   Max 1st sample XO time among all SL packets: 25.494467606
   TM values from -272 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-15T03:09:13.315.567.725 (XO: 6.315570166 - 6.315876167, t=37.2 degC)  Rank: 00079
   TM values: Max =   204  Min =  -264  Ave =    -4.44  Std =    20.99
   Min TM value:-264 at an offset of 118.938 usec -> 2021-07-15T03:09:13.315.686.662 (XO: 6.315689104)
   Max TM value:204 at an offset of 118.613 usec -> 2021-07-15T03:09:13.315.686.337 (XO: 6.315688779)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 121.325 usec -> 2021-07-15T03:09:13.315.689.050 (XO: 6.315691492)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 118.750 usec -> 2021-07-15T03:09:13.315.686.475 (XO: 6.315688917)

 - Block number: 2 starting at 2021-07-15T03:09:32.494.175.725 (XO: 25.494178605 - 25.494484606, t=37.2 degC)  Rank: 00079
   TM values: Max =   227  Min =  -272  Ave =    -4.36  Std =    30.57
   Min TM value:-272 at an offset of 190.763 usec -> 2021-07-15T03:09:32.494.366.488 (XO: 25.494369368)
   Max TM value:227 at an offset of 119.600 usec -> 2021-07-15T03:09:32.494.295.325 (XO: 25.494298206)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 175.313 usec -> 2021-07-15T03:09:32.494.351.038 (XO: 25.494353918)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 177.588 usec -> 2021-07-15T03:09:32.494.353.313 (XO: 25.494356193)
