-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 10:50:47 2021 UTC
-- File 20210708_235416_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.638579 to 41.499060
-each separately from 8s to 41s and from 498771us to 758869us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.119712 in 1 packets: 18
      13.739902 in 1 packets: 36
MIU packet time from 2021-07-08T23:54:27.638.579.000 to 2021-07-08T23:55:00.499.060.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.119712 in 1 packets: 18
      13.739902 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9387279 to 42.9386164
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999358 in 1 packets: 18
      13.9999527 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210708_235416_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5773079 to 10.5775969
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1196470  -- OK for SL --
   MIU time from 2021-07-08T23:54:27.638.579.000 to 2021-07-08T23:54:27.638.868.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.119712

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.6972439 to 29.6975329
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7398550  -- OK for SL --
   MIU time from 2021-07-08T23:54:46.758.580.000 to 2021-07-08T23:54:46.758.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.739902

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.4373879 to 43.4376769
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-08T23:55:00.498.771.000 to 2021-07-08T23:55:00.499.060.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.577307900 corrected by 40bit 10MHz counter of 9.938727900 obtained at 1Hz counter of 8. Result: 8.638582171
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-08T23:54:27.638.438.725
   Max 1st sample MIU time among all SL packets: 2021-07-08T23:55:00.498.919.725
   Min 1st sample XO time among all SL packets: 8.638441896
   Max 1st sample XO time among all SL packets: 41.498921921
   TM values from -256 to 219 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-08T23:54:27.638.438.725 (XO: 8.638441896 - 8.638747897, t=37.2 degC)  Rank: 00084
   TM values: Max =   209  Min =  -219  Ave =    -4.39  Std =    19.63
   Min TM value:-219 at an offset of 127.488 usec -> 2021-07-08T23:54:27.638.566.212 (XO: 8.638569384)
   Max TM value:209 at an offset of 130.175 usec -> 2021-07-08T23:54:27.638.568.900 (XO: 8.638572071)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 130.025 usec -> 2021-07-08T23:54:27.638.568.750 (XO: 8.638571921)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 114.588 usec -> 2021-07-08T23:54:27.638.553.312 (XO: 8.638556484)

 - Block number: 2 starting at 2021-07-08T23:54:46.758.439.725 (XO: 27.758442428 - 27.758748429, t=37.2 degC)  Rank: 00084
   TM values: Max =   210  Min =  -232  Ave =    -4.45  Std =    18.95
   Min TM value:-232 at an offset of 128.513 usec -> 2021-07-08T23:54:46.758.568.237 (XO: 27.758570941)
   Max TM value:210 at an offset of 131.138 usec -> 2021-07-08T23:54:46.758.570.862 (XO: 27.758573566)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 131.013 usec -> 2021-07-08T23:54:46.758.570.737 (XO: 27.758573441)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 128.288 usec -> 2021-07-08T23:54:46.758.568.012 (XO: 27.758570716)

 - Block number: 3 starting at 2021-07-08T23:55:00.498.630.725 (XO: 41.498632920 - 41.498938921, t=37.2 degC)  Rank: 00076
   TM values: Max =   219  Min =  -256  Ave =    -4.53  Std =    18.46
   Min TM value:-256 at an offset of 118.738 usec -> 2021-07-08T23:55:00.498.749.462 (XO: 41.498751658)
   Max TM value:219 at an offset of 118.438 usec -> 2021-07-08T23:55:00.498.749.162 (XO: 41.498751358)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 121.125 usec -> 2021-07-08T23:55:00.498.751.850 (XO: 41.498754046)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 118.538 usec -> 2021-07-08T23:55:00.498.749.262 (XO: 41.498751458)
