-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 04:15:37 2021 UTC
-- File 20210708_110616_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.735789 to 23.985911
-each separately from 11s to 23s and from 735789us to 985911us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.249544 in 1 packets: 18
MIU packet time from 2021-07-08T11:06:30.735.789.000 to 2021-07-08T11:06:42.985.911.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.249544 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.6170239 to 24.6169839
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999600 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210708_110616_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3528141 to 13.3531031
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.2495030  -- OK for SL --
   MIU time from 2021-07-08T11:06:30.735.789.000 to 2021-07-08T11:06:30.736.078.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.249544

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.6026061 to 25.6028951
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-08T11:06:42.985.622.000 to 2021-07-08T11:06:42.985.911.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.352814100 corrected by 40bit 10MHz counter of 12.617023900 obtained at 1Hz counter of 11. Result: 11.735792702
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-08T11:06:30.735.648.725
   Max 1st sample MIU time among all SL packets: 2021-07-08T11:06:42.985.770.725
   Min 1st sample XO time among all SL packets: 11.735652426
   Max 1st sample XO time among all SL packets: 23.985774178
   TM values from -252 to 221 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-08T11:06:30.735.648.725 (XO: 11.735652426 - 11.735958427, t=37.2 degC)  Rank: 00078
   TM values: Max =   221  Min =  -239  Ave =    -4.59  Std =    17.25
   Min TM value:-239 at an offset of 115.675 usec -> 2021-07-08T11:06:30.735.764.400 (XO: 11.735768102)
   Max TM value:221 at an offset of 115.388 usec -> 2021-07-08T11:06:30.735.764.112 (XO: 11.735767814)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 113.275 usec -> 2021-07-08T11:06:30.735.762.000 (XO: 11.735765702)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 115.500 usec -> 2021-07-08T11:06:30.735.764.225 (XO: 11.735767927)

 - Block number: 2 starting at 2021-07-08T11:06:42.985.481.725 (XO: 23.985485177 - 23.985791178, t=37.2 degC)  Rank: 00084
   TM values: Max =   215  Min =  -252  Ave =    -4.54  Std =    24.05
   Min TM value:-252 at an offset of 126.400 usec -> 2021-07-08T11:06:42.985.608.125 (XO: 23.985611578)
   Max TM value:215 at an offset of 126.100 usec -> 2021-07-08T11:06:42.985.607.825 (XO: 23.985611278)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 128.775 usec -> 2021-07-08T11:06:42.985.610.500 (XO: 23.985613953)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 155.851 usec -> 2021-07-08T11:06:42.985.637.575 (XO: 23.985641028)
