-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 02:57:49 2021 UTC
-- File 20210708_080207_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.163440 to 56.462057
-each separately from 11s to 56s and from 22484us to 462057us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.259386 in 1 packets: 18
      16.599080 in 1 packets: 36
      20.438995 in 1 packets: 54
MIU packet time from 2021-07-08T08:02:21.163.440.000 to 2021-07-08T08:03:06.462.057.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.259386 in 1 packets: 18
      16.599080 in 1 packets: 36
      20.438995 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4339681 to 57.4338162
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999729 in 1 packets: 18
      16.9999427 in 1 packets: 36
      19.9999325 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210708_080207_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.5974091 to 12.5976981
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2593590  -- OK for SL --
   MIU time from 2021-07-08T08:02:21.163.440.000 to 2021-07-08T08:02:21.163.729.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.259386

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.8570571 to 20.8573461
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5990230  -- OK for SL --
   MIU time from 2021-07-08T08:02:29.423.115.000 to 2021-07-08T08:02:29.423.404.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.599080

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4563691 to 37.4566581
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4389270  -- OK for SL --
   MIU time from 2021-07-08T08:02:46.022.484.000 to 2021-07-08T08:02:46.022.773.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.438995

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.8955851 to 57.8958741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-08T08:03:06.461.768.000 to 2021-07-08T08:03:06.462.057.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.597409100 corrected by 40bit 10MHz counter of 12.433968100 obtained at 1Hz counter of 11. Result: 11.163441556
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-08T08:02:21.163.299.725
   Max 1st sample MIU time among all SL packets: 2021-07-08T08:03:06.461.916.725
   Min 1st sample XO time among all SL packets: 11.163301280
   Max 1st sample XO time among all SL packets: 56.461919196
   TM values from -237 to 202 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-08T08:02:21.163.299.725 (XO: 11.163301280 - 11.163607281, t=37.2 degC)  Rank: 00087
   TM values: Max =   200  Min =  -237  Ave =    -4.55  Std =    21.17
   Min TM value:-237 at an offset of 118.175 usec -> 2021-07-08T08:02:21.163.417.900 (XO: 11.163419456)
   Max TM value:200 at an offset of 117.863 usec -> 2021-07-08T08:02:21.163.417.587 (XO: 11.163419143)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 120.575 usec -> 2021-07-08T08:02:21.163.420.300 (XO: 11.163421856)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 117.975 usec -> 2021-07-08T08:02:21.163.417.700 (XO: 11.163419256)

 - Block number: 2 starting at 2021-07-08T08:02:29.422.974.725 (XO: 19.422977221 - 19.423283222, t=37.2 degC)  Rank: 00077
   TM values: Max =   174  Min =  -225  Ave =    -4.50  Std =    20.93
   Min TM value:-225 at an offset of 116.350 usec -> 2021-07-08T08:02:29.423.091.075 (XO: 19.423093571)
   Max TM value:174 at an offset of 118.938 usec -> 2021-07-08T08:02:29.423.093.662 (XO: 19.423096159)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 118.750 usec -> 2021-07-08T08:02:29.423.093.475 (XO: 19.423095971)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 116.150 usec -> 2021-07-08T08:02:29.423.090.875 (XO: 19.423093371)

 - Block number: 3 starting at 2021-07-08T08:02:46.022.343.725 (XO: 36.022345201 - 36.022651202, t=37.2 degC)  Rank: 00081
   TM values: Max =   182  Min =  -229  Ave =    -4.49  Std =    18.64
   Min TM value:-229 at an offset of 106.300 usec -> 2021-07-08T08:02:46.022.450.025 (XO: 36.022451501)
   Max TM value:182 at an offset of 108.900 usec -> 2021-07-08T08:02:46.022.452.625 (XO: 36.022454101)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 108.700 usec -> 2021-07-08T08:02:46.022.452.425 (XO: 36.022453901)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 106.063 usec -> 2021-07-08T08:02:46.022.449.787 (XO: 36.022451264)

 - Block number: 4 starting at 2021-07-08T08:03:06.461.627.725 (XO: 56.461630195 - 56.461936196, t=37.2 degC)  Rank: 00077
   TM values: Max =   202  Min =  -231  Ave =    -4.42  Std =    21.38
   Min TM value:-231 at an offset of 109.700 usec -> 2021-07-08T08:03:06.461.737.425 (XO: 56.461739895)
   Max TM value:202 at an offset of 109.388 usec -> 2021-07-08T08:03:06.461.737.112 (XO: 56.461739582)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 112.050 usec -> 2021-07-08T08:03:06.461.739.775 (XO: 56.461742245)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 109.488 usec -> 2021-07-08T08:03:06.461.737.212 (XO: 56.461739682)
