-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 00:42:15 2021 UTC
-- File 20210707_151226_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.405593 to 24.575267
-each separately from 12s to 24s and from 405593us to 575267us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.169096 in 1 packets: 18
MIU packet time from 2021-07-07T15:12:41.405.593.000 to 2021-07-07T15:12:53.575.267.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.169096 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9176607 to 25.9176193
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999586 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210707_151226_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.3232542 to 14.3235432
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1690550  -- OK for SL --
   MIU time from 2021-07-07T15:12:41.405.593.000 to 2021-07-07T15:12:41.405.882.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.169096

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.4925982 to 26.4928872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-07T15:12:53.574.978.000 to 2021-07-07T15:12:53.575.267.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.323254200 corrected by 40bit 10MHz counter of 13.917660700 obtained at 1Hz counter of 12. Result: 12.405594879
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-07T15:12:41.405.452.725
   Max 1st sample MIU time among all SL packets: 2021-07-07T15:12:53.575.126.725
   Min 1st sample XO time among all SL packets: 12.405454604
   Max 1st sample XO time among all SL packets: 24.575129638
   TM values from -292 to 273 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-07T15:12:41.405.452.725 (XO: 12.405454604 - 12.405760605, t=37.2 degC)  Rank: 00084
   TM values: Max =   273  Min =  -292  Ave =    -4.48  Std =    27.41
   Min TM value:-292 at an offset of 115.950 usec -> 2021-07-07T15:12:41.405.568.675 (XO: 12.405570554)
   Max TM value:273 at an offset of 113.438 usec -> 2021-07-07T15:12:41.405.566.162 (XO: 12.405568041)
   Max dTM/dt [1/us]:2.68e+03 at an offset of 116.088 usec -> 2021-07-07T15:12:41.405.568.812 (XO: 12.405570691)
   Min dTM/dt [1/us]:-2.83e+03 at an offset of 113.525 usec -> 2021-07-07T15:12:41.405.566.250 (XO: 12.405568129)

 - Block number: 2 starting at 2021-07-07T15:12:53.574.837.725 (XO: 24.574840637 - 24.575146638, t=37.2 degC)  Rank: 00081
   TM values: Max =   240  Min =  -241  Ave =    -4.41  Std =    31.44
   Min TM value:-241 at an offset of 114.988 usec -> 2021-07-07T15:12:53.574.952.712 (XO: 24.574955625)
   Max TM value:240 at an offset of 117.788 usec -> 2021-07-07T15:12:53.574.955.512 (XO: 24.574958425)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 188.276 usec -> 2021-07-07T15:12:53.575.026.000 (XO: 24.575028913)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 114.888 usec -> 2021-07-07T15:12:53.574.952.612 (XO: 24.574955525)
