-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov  8 20:17:49 2021 UTC
-- File 20210707_045602_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.763382 to 12.763671
-each separately from 12s to 12s and from 763382us to 763671us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-07-07T04:56:17.763.382.000 to 2021-07-07T04:56:17.763.671.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9718838 to 13.9718838
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210707_045602_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.7352667 to 14.7355557
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-07T04:56:17.763.382.000 to 2021-07-07T04:56:17.763.671.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.735266700 corrected by 40bit 10MHz counter of 13.971883800 obtained at 1Hz counter of 12. Result: 12.763385648
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-07T04:56:17.763.241.725
   Max 1st sample MIU time among all SL packets: 2021-07-07T04:56:17.763.530.725
   Min 1st sample XO time among all SL packets: 12.763245373
   Max 1st sample XO time among all SL packets: 12.763534374
   TM values from -173 to 171 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-07T04:56:17.763.241.725 (XO: 12.763245373 - 12.763551374, t=37.2 degC)  Rank: 00069
   TM values: Max =   171  Min =  -173  Ave =    -4.51  Std =    58.88
   Min TM value:-173 at an offset of 123.663 usec -> 2021-07-07T04:56:17.763.365.387 (XO: 12.763369036)
   Max TM value:171 at an offset of 130.250 usec -> 2021-07-07T04:56:17.763.371.975 (XO: 12.763375623)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 130.225 usec -> 2021-07-07T04:56:17.763.371.950 (XO: 12.763375598)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 108.825 usec -> 2021-07-07T04:56:17.763.350.550 (XO: 12.763354198)
