-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Nov  8 17:00:18 2021 UTC
-- File 20210706_225434_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.472815 to 19.823005
-each separately from 2s to 19s and from 472815us to 823005us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.349612 in 1 packets: 18
MIU packet time from 2021-07-06T22:54:39.472.815.000 to 2021-07-06T22:54:56.823.005.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.349612 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8938839 to 20.8938233
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999394 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_225434_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3666998 to 4.3669888
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3495510  -- OK for SL --
   MIU time from 2021-07-06T22:54:39.472.815.000 to 2021-07-06T22:54:39.473.104.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.349612

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7165398 to 21.7168288
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T22:54:56.822.716.000 to 2021-07-06T22:54:56.823.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.366699800 corrected by 40bit 10MHz counter of 3.893883900 obtained at 1Hz counter of 2. Result: 2.472817602
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-06T22:54:39.472.674.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T22:54:56.822.864.725
   Min 1st sample XO time among all SL packets: 2.472677327
   Max 1st sample XO time among all SL packets: 19.822868270
   TM values from -246 to 235 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T22:54:39.472.674.725 (XO: 2.472677327 - 2.472983328, t=37.2 degC)  Rank: 00081
   TM values: Max =   235  Min =  -246  Ave =    -4.59  Std =    20.44
   Min TM value:-246 at an offset of 123.275 usec -> 2021-07-06T22:54:39.472.798.000 (XO: 2.472800602)
   Max TM value:235 at an offset of 125.900 usec -> 2021-07-06T22:54:39.472.800.625 (XO: 2.472803227)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 125.750 usec -> 2021-07-06T22:54:39.472.800.475 (XO: 2.472803077)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 123.138 usec -> 2021-07-06T22:54:39.472.797.862 (XO: 2.472800465)

 - Block number: 2 starting at 2021-07-06T22:54:56.822.575.725 (XO: 19.822579269 - 19.822885270, t=37.2 degC)  Rank: 00074
   TM values: Max =   212  Min =  -239  Ave =    -4.65  Std =    20.87
   Min TM value:-239 at an offset of 111.763 usec -> 2021-07-06T22:54:56.822.687.487 (XO: 19.822691031)
   Max TM value:212 at an offset of 114.500 usec -> 2021-07-06T22:54:56.822.690.225 (XO: 19.822693769)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 114.325 usec -> 2021-07-06T22:54:56.822.690.050 (XO: 19.822693594)
   Min dTM/dt [1/us]:-1.45e+03 at an offset of 111.625 usec -> 2021-07-06T22:54:56.822.687.350 (XO: 19.822690894)
