-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 12:12:44 2023 UTC
-- File 20210706_204958_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.492319 to 21.612508
-each separately from 9s to 21s and from 492319us to 612508us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.119611 in 1 packets: 18
MIU packet time from 2021-07-06T20:50:10.492.319.000 to 2021-07-06T20:50:22.612.508.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.119611 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4493188 to 22.4492756
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999568 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_204958_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.9416389 to 10.9419279
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1195670  -- OK for SL --
   MIU time from 2021-07-06T20:50:10.492.319.000 to 2021-07-06T20:50:10.492.608.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.119611

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.0614949 to 23.0617839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T20:50:22.612.219.000 to 2021-07-06T20:50:22.612.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.941638900 corrected by 40bit 10MHz counter of 10.449318800 obtained at 1Hz counter of 9. Result: 9.492321872
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-06T20:50:10.492.178.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T20:50:22.612.367.725
   Min 1st sample XO time among all SL packets: 9.492181597
   Max 1st sample XO time among all SL packets: 21.612370168
   TM values from -302 to 247 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T20:50:10.492.178.725 (XO: 9.492181597 - 9.492487598, t=37.2 degC)  Rank: 00083
   TM values: Max =   247  Min =  -284  Ave =    -4.43  Std =    24.85
   Min TM value:-284 at an offset of 116.963 usec -> 2021-07-06T20:50:10.492.295.687 (XO: 9.492298560)
   Max TM value:247 at an offset of 116.663 usec -> 2021-07-06T20:50:10.492.295.387 (XO: 9.492298260)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 119.338 usec -> 2021-07-06T20:50:10.492.298.062 (XO: 9.492300935)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 116.775 usec -> 2021-07-06T20:50:10.492.295.500 (XO: 9.492298372)

 - Block number: 2 starting at 2021-07-06T20:50:22.612.078.725 (XO: 21.612081167 - 21.612387168, t=37.2 degC)  Rank: 00076
   TM values: Max =   221  Min =  -302  Ave =    -4.47  Std =    23.78
   Min TM value:-302 at an offset of 118.975 usec -> 2021-07-06T20:50:22.612.197.700 (XO: 21.612200143)
   Max TM value:221 at an offset of 118.613 usec -> 2021-07-06T20:50:22.612.197.337 (XO: 21.612199780)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 121.275 usec -> 2021-07-06T20:50:22.612.200.000 (XO: 21.612202443)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 118.775 usec -> 2021-07-06T20:50:22.612.197.500 (XO: 21.612199943)
