-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 12:12:34 2023 UTC
-- File 20210706_203307_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.611482 to 30.951557
-each separately from 14s to 30s and from 381317us to 951557us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.769546 in 1 packets: 18
      4.569662 in 1 packets: 36
MIU packet time from 2021-07-06T20:33:24.611.482.000 to 2021-07-06T20:33:40.951.557.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.769546 in 1 packets: 18
      4.569662 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7830501 to 31.7829926
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999569 in 1 packets: 18
      3.9999856 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_203307_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.3945326 to 16.3948216
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7695030  -- OK for SL --
   MIU time from 2021-07-06T20:33:24.611.482.000 to 2021-07-06T20:33:24.611.771.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.769546

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.1643246 to 28.1646136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.5696470  -- OK for SL --
   MIU time from 2021-07-06T20:33:36.381.317.000 to 2021-07-06T20:33:36.381.606.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.569662

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.7342606 to 32.7345496
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T20:33:40.951.268.000 to 2021-07-06T20:33:40.951.557.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.394532600 corrected by 40bit 10MHz counter of 15.783050100 obtained at 1Hz counter of 14. Result: 14.611484701
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-06T20:33:24.611.341.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T20:33:40.951.416.725
   Min 1st sample XO time among all SL packets: 14.611344426
   Max 1st sample XO time among all SL packets: 30.951420245
   TM values from -297 to 271 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T20:33:24.611.341.725 (XO: 14.611344426 - 14.611650427, t=37.2 degC)  Rank: 00077
   TM values: Max =   223  Min =  -293  Ave =    -4.60  Std =    25.17
   Min TM value:-293 at an offset of 121.600 usec -> 2021-07-06T20:33:24.611.463.325 (XO: 14.611466026)
   Max TM value:223 at an offset of 124.200 usec -> 2021-07-06T20:33:24.611.465.925 (XO: 14.611468626)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 124.113 usec -> 2021-07-06T20:33:24.611.465.837 (XO: 14.611468539)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 121.375 usec -> 2021-07-06T20:33:24.611.463.100 (XO: 14.611465801)

 - Block number: 2 starting at 2021-07-06T20:33:36.381.176.725 (XO: 26.381178735 - 26.381484736, t=37.2 degC)  Rank: 00073
   TM values: Max =   271  Min =  -297  Ave =    -4.51  Std =    33.05
   Min TM value:-297 at an offset of 167.613 usec -> 2021-07-06T20:33:36.381.344.338 (XO: 26.381346348)
   Max TM value:271 at an offset of 109.075 usec -> 2021-07-06T20:33:36.381.285.800 (XO: 26.381287811)
   Max dTM/dt [1/us]:2.63e+03 at an offset of 108.950 usec -> 2021-07-06T20:33:36.381.285.675 (XO: 26.381287686)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 167.488 usec -> 2021-07-06T20:33:36.381.344.213 (XO: 26.381346223)

 - Block number: 3 starting at 2021-07-06T20:33:40.951.127.725 (XO: 30.951131244 - 30.951437245, t=37.2 degC)  Rank: 00072
   TM values: Max =   269  Min =  -297  Ave =    -4.49  Std =    32.83
   Min TM value:-297 at an offset of 136.138 usec -> 2021-07-06T20:33:40.951.263.863 (XO: 30.951267382)
   Max TM value:269 at an offset of 94.113 usec -> 2021-07-06T20:33:40.951.221.837 (XO: 30.951225357)
   Max dTM/dt [1/us]:2.78e+03 at an offset of 96.775 usec -> 2021-07-06T20:33:40.951.224.500 (XO: 30.951228020)
   Min dTM/dt [1/us]:-2.91e+03 at an offset of 94.225 usec -> 2021-07-06T20:33:40.951.221.950 (XO: 30.951225470)
