-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov  4 12:55:18 2021 UTC
-- File 20210706_080920_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.382453 to 20.942752
-each separately from 4s to 20s and from 382453us to 942752us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.559721 in 1 packets: 18
MIU packet time from 2021-07-06T08:09:22.382.453.000 to 2021-07-06T08:09:38.942.752.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.559721 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.6769774 to 21.6769208
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999434 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_080920_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.0594314 to 6.0597204
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5596630  -- OK for SL --
   MIU time from 2021-07-06T08:09:22.382.453.000 to 2021-07-06T08:09:22.382.742.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.559721

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.6193834 to 22.6196724
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T08:09:38.942.463.000 to 2021-07-06T08:09:38.942.752.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.059431400 corrected by 40bit 10MHz counter of 5.676977400 obtained at 1Hz counter of 4. Result: 4.382455339
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-06T08:09:22.382.312.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T08:09:38.942.611.725
   Min 1st sample XO time among all SL packets: 4.382315063
   Max 1st sample XO time among all SL packets: 20.942614718
   TM values from -256 to 242 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T08:09:22.382.312.725 (XO: 4.382315063 - 4.382621064, t=35.6 degC)  Rank: 00088
   TM values: Max =   197  Min =  -245  Ave =    -4.44  Std =    21.57
   Min TM value:-245 at an offset of 111.025 usec -> 2021-07-06T08:09:22.382.423.750 (XO: 4.382426088)
   Max TM value:197 at an offset of 126.325 usec -> 2021-07-06T08:09:22.382.439.050 (XO: 4.382441389)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 126.150 usec -> 2021-07-06T08:09:22.382.438.875 (XO: 4.382441214)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 123.463 usec -> 2021-07-06T08:09:22.382.436.187 (XO: 4.382438526)

 - Block number: 2 starting at 2021-07-06T08:09:38.942.322.725 (XO: 20.942325717 - 20.942631718, t=35.6 degC)  Rank: 00080
   TM values: Max =   242  Min =  -256  Ave =    -4.52  Std =    20.98
   Min TM value:-256 at an offset of 125.888 usec -> 2021-07-06T08:09:38.942.448.612 (XO: 20.942451605)
   Max TM value:242 at an offset of 125.600 usec -> 2021-07-06T08:09:38.942.448.325 (XO: 20.942451318)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 128.300 usec -> 2021-07-06T08:09:38.942.451.025 (XO: 20.942454018)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 125.700 usec -> 2021-07-06T08:09:38.942.448.425 (XO: 20.942451418)
