-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov  4 11:39:38 2021 UTC
-- File 20210706_033838_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.421288 to 89.321397
-each separately from 8s to 89s and from 11197us to 881569us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.269647 in 1 packets: 18
      19.319684 in 1 packets: 36
      9.559759 in 1 packets: 54
      19.309746 in 1 packets: 72
      23.439539 in 1 packets: 90
MIU packet time from 2021-07-06T03:38:44.421.288.000 to 2021-07-06T03:40:05.321.397.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.269647 in 1 packets: 18
      19.319684 in 1 packets: 36
      9.559759 in 1 packets: 54
      19.309746 in 1 packets: 72
      23.439539 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4624986 to 90.4622141
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999684 in 2 packets: 18, 54
      19.9999298 in 1 packets: 36
      18.9999334 in 1 packets: 72
      23.9999155 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 108 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_033838_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8837869 to 9.8840759
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2696150  -- OK for SL --
   MIU time from 2021-07-06T03:38:44.421.288.000 to 2021-07-06T03:38:44.421.577.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.269647

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1536909 to 19.1539799
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3196150  -- OK for SL --
   MIU time from 2021-07-06T03:38:53.691.224.000 to 2021-07-06T03:38:53.691.513.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.319684

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.4735949 to 38.4738839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5597270  -- OK for SL --
   MIU time from 2021-07-06T03:39:13.011.197.000 to 2021-07-06T03:39:13.011.486.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.559759

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.0336109 to 48.0338999
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3096790  -- OK for SL --
   MIU time from 2021-07-06T03:39:22.571.245.000 to 2021-07-06T03:39:22.571.534.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.309746

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.3435789 to 67.3438679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.4394550  -- OK for SL --
   MIU time from 2021-07-06T03:39:41.881.280.000 to 2021-07-06T03:39:41.881.569.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.439539

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 90.7833229 to 90.7836119
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T03:40:05.321.108.000 to 2021-07-06T03:40:05.321.397.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.883786900 corrected by 40bit 10MHz counter of 9.462498600 obtained at 1Hz counter of 8. Result: 8.421289817
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-06T03:38:44.421.147.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T03:40:05.321.256.725
   Min 1st sample XO time among all SL packets: 8.421149541
   Max 1st sample XO time among all SL packets: 89.321258649
   TM values from -261 to 229 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T03:38:44.421.147.725 (XO: 8.421149541 - 8.421455542, t=35.6 degC)  Rank: 00078
   TM values: Max =   214  Min =  -255  Ave =    -4.54  Std =    22.88
   Min TM value:-255 at an offset of 123.425 usec -> 2021-07-06T03:38:44.421.271.150 (XO: 8.421272967)
   Max TM value:214 at an offset of 126.088 usec -> 2021-07-06T03:38:44.421.273.812 (XO: 8.421275629)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 94.013 usec -> 2021-07-06T03:38:44.421.241.737 (XO: 8.421243554)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 96.263 usec -> 2021-07-06T03:38:44.421.243.987 (XO: 8.421245804)

 - Block number: 2 starting at 2021-07-06T03:38:53.691.083.725 (XO: 17.691086044 - 17.691392045, t=35.6 degC)  Rank: 00083
   TM values: Max =   216  Min =  -261  Ave =    -4.51  Std =    18.78
   Min TM value:-261 at an offset of 123.075 usec -> 2021-07-06T03:38:53.691.206.800 (XO: 17.691209119)
   Max TM value:216 at an offset of 122.825 usec -> 2021-07-06T03:38:53.691.206.550 (XO: 17.691208869)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 120.600 usec -> 2021-07-06T03:38:53.691.204.325 (XO: 17.691206644)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 122.950 usec -> 2021-07-06T03:38:53.691.206.675 (XO: 17.691208994)

 - Block number: 3 starting at 2021-07-06T03:39:13.011.056.725 (XO: 37.011057864 - 37.011363865, t=35.6 degC)  Rank: 00077
   TM values: Max =   229  Min =  -247  Ave =    -4.36  Std =    18.91
   Min TM value:-247 at an offset of 110.800 usec -> 2021-07-06T03:39:13.011.167.525 (XO: 37.011168664)
   Max TM value:229 at an offset of 113.525 usec -> 2021-07-06T03:39:13.011.170.250 (XO: 37.011171389)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 113.425 usec -> 2021-07-06T03:39:13.011.170.150 (XO: 37.011171289)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 110.625 usec -> 2021-07-06T03:39:13.011.167.350 (XO: 37.011168489)

 - Block number: 4 starting at 2021-07-06T03:39:22.571.104.725 (XO: 46.571107424 - 46.571413425, t=35.6 degC)  Rank: 00078
   TM values: Max =   214  Min =  -248  Ave =    -4.51  Std =    18.37
   Min TM value:-248 at an offset of 122.038 usec -> 2021-07-06T03:39:22.571.226.762 (XO: 46.571229462)
   Max TM value:214 at an offset of 124.700 usec -> 2021-07-06T03:39:22.571.229.425 (XO: 46.571232124)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 124.488 usec -> 2021-07-06T03:39:22.571.229.212 (XO: 46.571231912)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 121.900 usec -> 2021-07-06T03:39:22.571.226.625 (XO: 46.571229324)

 - Block number: 5 starting at 2021-07-06T03:39:41.881.139.725 (XO: 65.881143109 - 65.881449110, t=35.6 degC)  Rank: 00078
   TM values: Max =   208  Min =  -257  Ave =    -4.50  Std =    20.79
   Min TM value:-257 at an offset of 121.150 usec -> 2021-07-06T03:39:41.881.260.875 (XO: 65.881264259)
   Max TM value:208 at an offset of 123.775 usec -> 2021-07-06T03:39:41.881.263.500 (XO: 65.881266884)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 123.663 usec -> 2021-07-06T03:39:41.881.263.387 (XO: 65.881266772)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 120.925 usec -> 2021-07-06T03:39:41.881.260.650 (XO: 65.881264034)

 - Block number: 6 starting at 2021-07-06T03:40:05.320.967.725 (XO: 89.320969648 - 89.321275649, t=35.6 degC)  Rank: 00083
   TM values: Max =   213  Min =  -247  Ave =    -4.45  Std =    20.82
   Min TM value:-247 at an offset of 82.863 usec -> 2021-07-06T03:40:05.321.050.587 (XO: 89.321052511)
   Max TM value:213 at an offset of 98.425 usec -> 2021-07-06T03:40:05.321.066.150 (XO: 89.321068074)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 98.263 usec -> 2021-07-06T03:40:05.321.065.987 (XO: 89.321067911)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 82.688 usec -> 2021-07-06T03:40:05.321.050.412 (XO: 89.321052336)
