-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov  4 10:28:45 2021 UTC
-- File 20210706_021917_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.835512 to 41.055840
-each separately from 2s to 41s and from 55551us to 985753us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.149663 in 1 packets: 18
      15.069798 in 1 packets: 36
MIU packet time from 2021-07-06T02:19:18.835.512.000 to 2021-07-06T02:19:57.055.840.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.149663 in 1 packets: 18
      15.069798 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9498595 to 42.9497231
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999196 in 1 packets: 18
      15.9999440 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210706_021917_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.7853712 to 4.7856602
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1495830  -- OK for SL --
   MIU time from 2021-07-06T02:19:18.835.512.000 to 2021-07-06T02:19:18.835.801.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.149663

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.9352432 to 27.9355322
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0697430  -- OK for SL --
   MIU time from 2021-07-06T02:19:41.985.464.000 to 2021-07-06T02:19:41.985.753.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.069798

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0052752 to 43.0055642
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-06T02:19:57.055.551.000 to 2021-07-06T02:19:57.055.840.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.785371200 corrected by 40bit 10MHz counter of 3.949859500 obtained at 1Hz counter of 2. Result: 2.835514624
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-06T02:19:18.835.371.725
   Max 1st sample MIU time among all SL packets: 2021-07-06T02:19:57.055.699.725
   Min 1st sample XO time among all SL packets: 2.835374349
   Max 1st sample XO time among all SL packets: 41.055701020
   TM values from -249 to 216 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-06T02:19:18.835.371.725 (XO: 2.835374349 - 2.835680350, t=35.6 degC)  Rank: 00083
   TM values: Max =   215  Min =  -234  Ave =    -4.54  Std =    18.51
   Min TM value:-234 at an offset of 108.413 usec -> 2021-07-06T02:19:18.835.480.137 (XO: 2.835482762)
   Max TM value:215 at an offset of 110.988 usec -> 2021-07-06T02:19:18.835.482.712 (XO: 2.835485337)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 110.825 usec -> 2021-07-06T02:19:18.835.482.550 (XO: 2.835485174)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 108.188 usec -> 2021-07-06T02:19:18.835.479.912 (XO: 2.835482537)

 - Block number: 2 starting at 2021-07-06T02:19:41.985.323.725 (XO: 25.985327274 - 25.985633275, t=35.6 degC)  Rank: 00081
   TM values: Max =   216  Min =  -249  Ave =    -4.54  Std =    28.77
   Min TM value:-249 at an offset of 122.100 usec -> 2021-07-06T02:19:41.985.445.825 (XO: 25.985449374)
   Max TM value:216 at an offset of 124.750 usec -> 2021-07-06T02:19:41.985.448.475 (XO: 25.985452024)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 189.951 usec -> 2021-07-06T02:19:41.985.513.675 (XO: 25.985517224)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 121.963 usec -> 2021-07-06T02:19:41.985.445.687 (XO: 25.985449237)

 - Block number: 3 starting at 2021-07-06T02:19:57.055.410.725 (XO: 41.055412019 - 41.055718020, t=35.6 degC)  Rank: 00080
   TM values: Max =   208  Min =  -220  Ave =    -4.55  Std =    18.77
   Min TM value:-220 at an offset of 108.338 usec -> 2021-07-06T02:19:57.055.519.062 (XO: 41.055520357)
   Max TM value:208 at an offset of 111.050 usec -> 2021-07-06T02:19:57.055.521.775 (XO: 41.055523069)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 110.888 usec -> 2021-07-06T02:19:57.055.521.612 (XO: 41.055522907)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 108.113 usec -> 2021-07-06T02:19:57.055.518.837 (XO: 41.055520132)
