-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov  4 08:43:30 2021 UTC
-- File 20210705_231424_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.793377 to 25.533450
-each separately from 9s to 25s and from 533161us to 793666us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.739495 in 1 packets: 18
MIU packet time from 2021-07-05T23:14:31.793.377.000 to 2021-07-05T23:14:47.533.450.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.739495 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8954783 to 26.8954222
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999439 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210705_231424_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6888552 to 11.6891442
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7394390  -- OK for SL --
   MIU time from 2021-07-05T23:14:31.793.377.000 to 2021-07-05T23:14:31.793.666.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.739495

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4285832 to 27.4288722
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-05T23:14:47.533.161.000 to 2021-07-05T23:14:47.533.450.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.688855200 corrected by 40bit 10MHz counter of 10.895478300 obtained at 1Hz counter of 9. Result: 9.793379677
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-05T23:14:31.793.236.725
   Max 1st sample MIU time among all SL packets: 2021-07-05T23:14:47.533.309.725
   Min 1st sample XO time among all SL packets: 9.793239401
   Max 1st sample XO time among all SL packets: 25.533311592
   TM values from -286 to 244 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-05T23:14:31.793.236.725 (XO: 9.793239401 - 9.793545402, t=35.6 degC)  Rank: 00083
   TM values: Max =   244  Min =  -268  Ave =    -4.55  Std =    25.23
   Min TM value:-268 at an offset of 127.700 usec -> 2021-07-05T23:14:31.793.364.425 (XO: 9.793367102)
   Max TM value:244 at an offset of 130.425 usec -> 2021-07-05T23:14:31.793.367.150 (XO: 9.793369827)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 130.313 usec -> 2021-07-05T23:14:31.793.367.037 (XO: 9.793369714)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 127.550 usec -> 2021-07-05T23:14:31.793.364.275 (XO: 9.793366952)

 - Block number: 2 starting at 2021-07-05T23:14:47.533.020.725 (XO: 25.533022591 - 25.533328592, t=35.6 degC)  Rank: 00087
   TM values: Max =   231  Min =  -286  Ave =    -4.52  Std =    21.63
   Min TM value:-286 at an offset of 129.638 usec -> 2021-07-05T23:14:47.533.150.362 (XO: 25.533152229)
   Max TM value:231 at an offset of 132.325 usec -> 2021-07-05T23:14:47.533.153.050 (XO: 25.533154916)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 132.200 usec -> 2021-07-05T23:14:47.533.152.925 (XO: 25.533154791)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 129.475 usec -> 2021-07-05T23:14:47.533.150.200 (XO: 25.533152066)
