-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 11:30:02 2023 UTC
-- File 20210705_161524_1229.dat, 14246  packets found.
APID:
      1229 (EW) in 14246 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 14246 packets (all packets)
Total number of packets: 
      36720 in 14246 packets (all packets)
Packet No from 1 to 14246
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 1.283351 to 5.455370
-each separately from 1s to 5s and from 283351us to 545707us.
   Differences in neigboring packets: 
      0.000000 in 14243 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1.262356 in 1 packets: 12036
      2.909663 in 1 packets: 13668
MIU packet time from 2021-07-05T16:15:24.283.351.000 to 2021-07-05T16:15:28.455.370.000
   Differences in neigboring packets: 
      0.000000 in 14243 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1.262356 in 1 packets: 12036
      2.909663 in 1 packets: 13668
MIU orbit number: 
      60000 in 14246 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 14246 packets (all packets)
MIU orbit information / number of events: 
      0 in 14246 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 14246 packets (all packets)
Instrument anomalies: 
      0 (OK) in 14246 packets (all packets)
FPGA configuration version: 
      211 in 14246 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.3525449 to 7.3525307
   Differences in neigboring packets: 
      0.0000000 in 14243 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      0.9999964 in 1 packets: 12036
      2.9999894 in 1 packets: 13668
Temperature FPGA: 
      05 -> 48.5 degC in 12614 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 1632 packets: 12037, 12038, 12039, 12040, 12041, 12042, 12043, 12044, 12045, 12046, 12047, 12048, 12049, 12050, 12051, 12052, 12053, 12054, ...
Temperature XO: 
      C0 -> 35.6 degC in 14246 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 14246 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 14246 packets (all packets)
Number of external events: 
      1 in 12036 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1632 packets: 12037, 12038, 12039, 12040, 12041, 12042, 12043, 12044, 12045, 12046, 12047, 12048, 12049, 12050, 12051, 12052, 12053, 12054, ...
      3 in 578 packets: 13669, 13670, 13671, 13672, 13673, 13674, 13675, 13676, 13677, 13678, 13679, 13680, 13681, 13682, 13683, 13684, 13685, 13686, ...
Number of generated alerts: 
      1 in 12036 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1632 packets: 12037, 12038, 12039, 12040, 12041, 12042, 12043, 12044, 12045, 12046, 12047, 12048, 12049, 12050, 12051, 12052, 12053, 12054, ...
      3 in 578 packets: 13669, 13670, 13671, 13672, 13673, 13674, 13675, 13676, 13677, 13678, 13679, 13680, 13681, 13682, 13683, 13684, 13685, 13686, ...
Survey memory error: 
      0 in 14246 packets (all packets)
Event memory error: 
      0 in 14246 packets (all packets)
Configuration error: 
      0 in 14246 packets (all packets)
SDRAM CRC error: 
      0 in 14246 packets (all packets)
Radio gain: 
      0 in 14246 packets (all packets)
Radio bandwidth: 
      0 in 14246 packets (all packets)
Digital gain: 
      0 in 14246 packets (all packets)
Analog gain: 
      0 in 14246 packets (all packets)

EVENT DATA READER
-- File 20210705_161524_1229.dat, 69 EW pages found. Extra packets: 170.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      1 in 12036 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 1632 packets: 12037, 12038, 12039, 12040, 12041, 12042, 12043, 12044, 12045, 12046, 12047, 12048, 12049, 12050, 12051, 12052, 12053, 12054, ...
      3 in 408 packets: 13669, 13670, 13671, 13672, 13673, 13674, 13675, 13676, 13677, 13678, 13679, 13680, 13681, 13682, 13683, 13684, 13685, 13686, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 3.5838768 to 7.9223278
      Difference between the neighboring packets: 
      0.0000170 in 14073 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1.2346250 in 1 packets: 12036
      2.8645850 in 1 packets: 13668
   Memory page number from 0 to 68
      Difference between the neighboring packets: 
      0 in 14007 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 68 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 2 to 60
      Difference between the neighboring packets: 
      0 in 14007 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 66 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -7 in 1 packets: 12036
      -12 in 1 packets: 13668
   MIU event flag 
      1 in 14076 packets (all packets)
   Time triggered event flag 
      0 in 14076 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2021-07-05T16:15:24.227.862.825
   Min EW header abs. time found among all events            : 2021-07-05T16:15:24.283.351.000
   Max EW event (1st sample) abs. time found among all events: 2021-07-05T16:15:28.399.881.825
   Max EW header abs. time found among all events            : 2021-07-05T16:15:28.455.370.000
   TM values from -289 to 274 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12036
    Page no. 2:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 59  ordered packets: 12036
   TM values from -269 to 274 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=1 C1MHz=0.283351
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12035 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 3.5838768
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2021-07-05T16:15:24.283.351.000
   Event 1st sample abs. time : 2021-07-05T16:15:24.227.862.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 2 total number of packets: 1632
    Page no. 53:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 8  ordered packets: 1632
   TM values from -258 to 237 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=2 C1MHz=0.545707
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 1631 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 5.0230968
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2021-07-05T16:15:25.545.707.000
   Event 1st sample abs. time : 2021-07-05T16:15:25.490.218.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000

 Event no. 3 total number of packets: 408
    Page no. 48:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 2  ordered packets: 408
   TM values from -289 to 210 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=5 C1MHz=0.455370
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 407 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 7.9154088
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2021-07-05T16:15:28.455.370.000
   Event 1st sample abs. time : 2021-07-05T16:15:28.399.881.825)
   --- Rel. time correction failed for 1 points and 0 XO calibration points.
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):            -333.000000175
   Event data descriptor XO time (reconstructed seconds from XP start)    :            -333.000000000
   Sampling frequency correction factor for this event:     1.000000000
--
Extremes in separate events:

 - Event number: 1 from 2021-07-05T16:15:24.227.862.825 to 2021-07-05T16:15:24.432.474.825 (XO: -333.000000175 - -332.795388175)
   Min TM value:-269 at an offset of 51554.312 usec -> 2021-07-05T16:15:24.279.417.137 (XO: -332.948445863)
   Max TM value:274 at an offset of 121497.312 usec -> 2021-07-05T16:15:24.349.360.137 (XO: -332.878502862)
   Max dTM/dt [1/us]:2.73e+03 at an offset of 41570.150 usec -> 2021-07-05T16:15:24.269.432.975 (XO: -332.958430025)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 61552.988 usec -> 2021-07-05T16:15:24.289.415.812 (XO: -332.938447188)
   Med: -4.00000   Ave: -4.4204775   Std: 10.300626

 - Event number: 2 from 2021-07-05T16:15:25.490.218.825 to 2021-07-05T16:15:25.517.962.825 (XO: -333.000000175 - -332.972256175)
   Min TM value:-258 at an offset of 12389.150 usec -> 2021-07-05T16:15:25.502.607.975 (XO: -332.987611025)
   Max TM value:237 at an offset of 22344.975 usec -> 2021-07-05T16:15:25.512.563.800 (XO: -332.977655200)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 22344.800 usec -> 2021-07-05T16:15:25.512.563.625 (XO: -332.977655375)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 12388.975 usec -> 2021-07-05T16:15:25.502.607.800 (XO: -332.987611200)
   Med: -4.00000   Ave: -4.3786305   Std: 9.7300318

 - Event number: 3 from 2021-07-05T16:15:28.399.881.825 to 2021-07-05T16:15:28.406.817.825 (XO: -333.000000175 - -332.993064175)
   Min TM value:-289 at an offset of 107.675 usec -> 2021-07-05T16:15:28.399.989.500 (XO: -332.999892500)
   Max TM value:210 at an offset of 138.312 usec -> 2021-07-05T16:15:28.400.020.137 (XO: -332.999861863)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 138.162 usec -> 2021-07-05T16:15:28.400.019.987 (XO: -332.999862013)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 107.562 usec -> 2021-07-05T16:15:28.399.989.387 (XO: -332.999892613)
   Med: -4.00000   Ave: -4.3758705   Std: 10.862392
