-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 30 14:19:15 2021 UTC
-- File 20210705_121347_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833944 in 5 packets: 3, 6, 10, 14, 17
      0.833941 in 2 packets: 7, 13
MIU packet time from 2021-07-05T12:13:46.000.000.000 to 2021-07-05T12:14:00.177.029.000
   Differences in neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833944 in 5 packets: 3, 6, 10, 14, 17
      0.833941 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9237646 to 15.9237151
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 9 packets: 2, 3, 5, 6, 8, 10, 12, 15, 16
      0.9999964 in 5 packets: 4, 9, 11, 14, 17
   XO PPM: 
      -3.5 in 9 packets: 2, 3, 5, 6, 8, 10, 12, 15, 16
      -3.6 in 5 packets: 4, 9, 11, 14, 17
Temperature FPGA: 
      05 -> 48.5 degC in 11 packets: 1, 2, 6, 10, 11, 12, 13, 14, 15, 17, 18
      06 -> 47.2 degC in 5 packets: 3, 5, 7, 8, 9
      04 -> 49.8 degC in 2 packets: 4, 16
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 14 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15
      2 in 3 packets: 16, 17, 18
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 14 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15
      2 in 3 packets: 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210705_121347_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9237647 to 16.1007447
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-05T12:13:46.000.000.000 to 2021-07-05T12:14:00.177.029.000
      Difference between neigboring packets: 
      0.833940 in 1 packets: 1
      0.833943 in 9 packets: 2, 4, 5, 8, 9, 11, 12, 15, 16
      0.833944 in 5 packets: 3, 6, 10, 14, 17
      0.833941 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.923764700 corrected by 40bit 10MHz counter of 1.923764600 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2021-07-05T12:13:45.999.999.925
   Max 1st sample MIU time among all SR packets: 2021-07-05T12:14:00.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177030162
   TM values from -121 to 105 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2021-07-05T12:13:45.999.999.925 (XO: 0.000000025 - 0.000017025, t=35.6 degC)
   TM values: Max =    18  Min =   -25  Ave =    -4.46  Std =     7.93

 - Packet number: 2 starting at 2021-07-05T12:13:46.833.939.925 (XO: 0.833942944 - 0.833959944, t=35.6 degC)
   TM values: Max =     8  Min =   -19  Ave =    -4.24  Std =     4.70

 - Packet number: 3 starting at 2021-07-05T12:13:47.667.882.925 (XO: 1.667885863 - 1.667902863, t=35.6 degC)
   TM values: Max =    16  Min =   -27  Ave =    -4.47  Std =     7.61

 - Packet number: 4 starting at 2021-07-05T12:13:48.501.826.925 (XO: 2.501828832 - 2.501845832, t=35.6 degC)
   TM values: Max =    15  Min =   -25  Ave =    -4.26  Std =     6.43

 - Packet number: 5 starting at 2021-07-05T12:13:49.335.769.925 (XO: 3.335771800 - 3.335788800, t=35.6 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.66  Std =     7.51

 - Packet number: 6 starting at 2021-07-05T12:13:50.169.712.925 (XO: 4.169714719 - 4.169731719, t=35.6 degC)
   TM values: Max =   105  Min =  -121  Ave =    -4.41  Std =    23.61

 - Packet number: 7 starting at 2021-07-05T12:13:51.003.656.925 (XO: 5.003657638 - 5.003674638, t=35.6 degC)
   TM values: Max =    12  Min =   -22  Ave =    -4.25  Std =     6.29

 - Packet number: 8 starting at 2021-07-05T12:13:51.837.597.925 (XO: 5.837600557 - 5.837617557, t=35.6 degC)
   TM values: Max =    20  Min =   -28  Ave =    -4.50  Std =     9.33

 - Packet number: 9 starting at 2021-07-05T12:13:52.671.540.925 (XO: 6.671543543 - 6.671560543, t=35.6 degC)
   TM values: Max =    18  Min =   -26  Ave =    -4.57  Std =     7.77

 - Packet number: 10 starting at 2021-07-05T12:13:53.505.483.925 (XO: 7.505486494 - 7.505503494, t=35.6 degC)
   TM values: Max =    17  Min =   -24  Ave =    -4.41  Std =     7.27

 - Packet number: 11 starting at 2021-07-05T12:13:54.339.427.925 (XO: 8.339429447 - 8.339446447, t=35.6 degC)
   TM values: Max =    11  Min =   -18  Ave =    -4.43  Std =     5.68

 - Packet number: 12 starting at 2021-07-05T12:13:55.173.370.925 (XO: 9.173372432 - 9.173389432, t=35.6 degC)
   TM values: Max =    26  Min =   -36  Ave =    -4.49  Std =    13.12

 - Packet number: 13 starting at 2021-07-05T12:13:56.007.313.925 (XO: 10.007315351 - 10.007332351, t=35.6 degC)
   TM values: Max =    18  Min =   -28  Ave =    -4.39  Std =     8.85

 - Packet number: 14 starting at 2021-07-05T12:13:56.841.254.925 (XO: 10.841258354 - 10.841275354, t=35.6 degC)
   TM values: Max =    25  Min =   -34  Ave =    -4.36  Std =    11.38

 - Packet number: 15 starting at 2021-07-05T12:13:57.675.198.925 (XO: 11.675201288 - 11.675218288, t=35.6 degC)
   TM values: Max =    14  Min =   -23  Ave =    -4.40  Std =     6.73

 - Packet number: 16 starting at 2021-07-05T12:13:58.509.141.925 (XO: 12.509144207 - 12.509161207, t=35.6 degC)
   TM values: Max =    18  Min =   -28  Ave =    -4.54  Std =    10.22

 - Packet number: 17 starting at 2021-07-05T12:13:59.343.084.925 (XO: 13.343087160 - 13.343104160, t=35.6 degC)
   TM values: Max =    11  Min =   -21  Ave =    -4.37  Std =     6.35

 - Packet number: 18 starting at 2021-07-05T12:14:00.177.028.925 (XO: 14.177030162 - 14.177047162, t=35.6 degC)
   TM values: Max =    61  Min =   -70  Ave =    -4.50  Std =    17.65
