-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 30 04:54:09 2021 UTC
-- File 20210704_105507_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.999604 to 17.169465
-each separately from 7s to 17s and from 169176us to 999893us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.169283 in 1 packets: 18
MIU packet time from 2021-07-04T10:55:13.999.604.000 to 2021-07-04T10:55:23.169.465.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.169283 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.6943634 to 18.6943268
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999634 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210704_105507_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6939673 to 9.6942563
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1692470  -- OK for SL --
   MIU time from 2021-07-04T10:55:13.999.604.000 to 2021-07-04T10:55:13.999.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.169283

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.8635033 to 18.8637923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-04T10:55:23.169.176.000 to 2021-07-04T10:55:23.169.465.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.693967300 corrected by 40bit 10MHz counter of 8.694363400 obtained at 1Hz counter of 7. Result: 7.999607499
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-04T10:55:13.999.463.725
   Max 1st sample MIU time among all SL packets: 2021-07-04T10:55:23.169.324.725
   Min 1st sample XO time among all SL packets: 7.999467223
   Max 1st sample XO time among all SL packets: 17.169325835
   TM values from -244 to 272 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-04T10:55:13.999.463.725 (XO: 7.999467223 - 7.999773224, t=35.6 degC)  Rank: 00082
   TM values: Max =   230  Min =  -244  Ave =    -4.43  Std =    23.89
   Min TM value:-244 at an offset of 124.088 usec -> 2021-07-04T10:55:13.999.587.812 (XO: 7.999591311)
   Max TM value:230 at an offset of 126.863 usec -> 2021-07-04T10:55:13.999.590.587 (XO: 7.999594086)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 126.688 usec -> 2021-07-04T10:55:13.999.590.412 (XO: 7.999593911)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 111.625 usec -> 2021-07-04T10:55:13.999.575.350 (XO: 7.999578848)

 - Block number: 2 starting at 2021-07-04T10:55:23.169.035.725 (XO: 17.169036834 - 17.169342835, t=35.6 degC)  Rank: 00083
   TM values: Max =   272  Min =  -236  Ave =    -4.45  Std =    17.78
   Min TM value:-236 at an offset of 99.563 usec -> 2021-07-04T10:55:23.169.135.287 (XO: 17.169136396)
   Max TM value:272 at an offset of 102.275 usec -> 2021-07-04T10:55:23.169.138.000 (XO: 17.169139109)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 102.150 usec -> 2021-07-04T10:55:23.169.137.875 (XO: 17.169138984)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 99.388 usec -> 2021-07-04T10:55:23.169.135.112 (XO: 17.169136221)
