-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Oct 30 04:04:35 2021 UTC
-- File 20210704_094625_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.379011 to 22.879036
-each separately from 7s to 22s and from 379011us to 879036us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.499447 in 1 packets: 18
MIU packet time from 2021-07-04T09:46:31.379.011.000 to 2021-07-04T09:46:46.879.036.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.499447 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8460989 to 23.8460435
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999446 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210704_094625_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2251102 to 9.2253992
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4993910  -- OK for SL --
   MIU time from 2021-07-04T09:46:31.379.011.000 to 2021-07-04T09:46:31.379.300.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.499447

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7247902 to 24.7250792
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-04T09:46:46.878.747.000 to 2021-07-04T09:46:46.879.036.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.225110200 corrected by 40bit 10MHz counter of 8.846098900 obtained at 1Hz counter of 7. Result: 7.379012740
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2021-07-04T09:46:31.378.870.725
   Max 1st sample MIU time among all SL packets: 2021-07-04T09:46:46.878.895.725
   Min 1st sample XO time among all SL packets: 7.378872465
   Max 1st sample XO time among all SL packets: 22.878898677
   TM values from -271 to 266 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-04T09:46:31.378.870.725 (XO: 7.378872465 - 7.379178466, t=35.6 degC)  Rank: 00083
   TM values: Max =   266  Min =  -271  Ave =    -4.47  Std =    27.26
   Min TM value:-271 at an offset of 111.150 usec -> 2021-07-04T09:46:31.378.981.875 (XO: 7.378983615)
   Max TM value:266 at an offset of 151.588 usec -> 2021-07-04T09:46:31.379.022.313 (XO: 7.379024053)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 151.463 usec -> 2021-07-04T09:46:31.379.022.188 (XO: 7.379023928)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 153.713 usec -> 2021-07-04T09:46:31.379.024.438 (XO: 7.379026178)

 - Block number: 2 starting at 2021-07-04T09:46:46.878.606.725 (XO: 22.878609676 - 22.878915677, t=35.6 degC)  Rank: 00079
   TM values: Max =   240  Min =  -267  Ave =    -4.51  Std =    28.44
   Min TM value:-267 at an offset of 176.076 usec -> 2021-07-04T09:46:46.878.782.800 (XO: 22.878785752)
   Max TM value:240 at an offset of 123.038 usec -> 2021-07-04T09:46:46.878.729.762 (XO: 22.878732714)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 122.913 usec -> 2021-07-04T09:46:46.878.729.637 (XO: 22.878732589)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 108.525 usec -> 2021-07-04T09:46:46.878.715.250 (XO: 22.878718201)
