-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 10:38:30 2023 UTC
-- File 20210703_112313_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.720790 to 29.458907
-each separately from 10s to 29s and from 458618us to 721079us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.737539 in 1 packets: 18
MIU packet time from 2021-07-03T11:23:22.720.790.000 to 2021-07-03T11:23:41.458.907.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.737539 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9439010 to 30.9438325
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999315 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210703_112313_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.6646908 to 12.6649798
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7374710  -- OK for SL --
   MIU time from 2021-07-03T11:23:22.720.790.000 to 2021-07-03T11:23:22.721.079.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.737539

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4024508 to 31.4027398
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-03T11:23:41.458.618.000 to 2021-07-03T11:23:41.458.907.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.664690800 corrected by 40bit 10MHz counter of 11.943901000 obtained at 1Hz counter of 10. Result: 10.720792395
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-03T11:23:22.720.649.725
   Max 1st sample MIU time among all SL packets: 2021-07-03T11:23:41.458.766.725
   Min 1st sample XO time among all SL packets: 10.720652119
   Max 1st sample XO time among all SL packets: 29.458768677
   TM values from -255 to 212 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-03T11:23:22.720.649.725 (XO: 10.720652119 - 10.720958120, t=35.6 degC)  Rank: 00084
   TM values: Max =   212  Min =  -255  Ave =    -4.54  Std =    21.62
   Min TM value:-255 at an offset of 108.300 usec -> 2021-07-03T11:23:22.720.758.025 (XO: 10.720760420)
   Max TM value:212 at an offset of 110.950 usec -> 2021-07-03T11:23:22.720.760.675 (XO: 10.720763070)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 110.788 usec -> 2021-07-03T11:23:22.720.760.512 (XO: 10.720762907)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 108.113 usec -> 2021-07-03T11:23:22.720.757.837 (XO: 10.720760232)

 - Block number: 2 starting at 2021-07-03T11:23:41.458.477.725 (XO: 29.458479676 - 29.458785677, t=35.6 degC)  Rank: 00080
   TM values: Max =   204  Min =  -240  Ave =    -4.50  Std =    18.40
   Min TM value:-240 at an offset of 120.863 usec -> 2021-07-03T11:23:41.458.598.587 (XO: 29.458600538)
   Max TM value:204 at an offset of 123.613 usec -> 2021-07-03T11:23:41.458.601.337 (XO: 29.458603288)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 123.438 usec -> 2021-07-03T11:23:41.458.601.162 (XO: 29.458603113)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 120.725 usec -> 2021-07-03T11:23:41.458.598.450 (XO: 29.458600401)
