-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 29 21:36:16 2021 UTC
-- File 20210703_110931_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.004110 to 9.004399
-each separately from 9s to 9s and from 4110us to 4399us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-07-03T11:09:39.004.110.000 to 2021-07-03T11:09:39.004.399.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4464130 to 10.4464130
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210703_110931_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.4505244 to 10.4508134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-03T11:09:39.004.110.000 to 2021-07-03T11:09:39.004.399.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.450524400 corrected by 40bit 10MHz counter of 10.446413000 obtained at 1Hz counter of 9. Result: 9.004111415
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-03T11:09:39.003.969.725
   Max 1st sample MIU time among all SL packets: 2021-07-03T11:09:39.004.258.725
   Min 1st sample XO time among all SL packets: 9.003971139
   Max 1st sample XO time among all SL packets: 9.004260140
   TM values from -242 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-03T11:09:39.003.969.725 (XO: 9.003971139 - 9.004277140, t=35.6 degC)  Rank: 00079
   TM values: Max =   232  Min =  -242  Ave =    -4.39  Std =    19.51
   Min TM value:-242 at an offset of 36.663 usec -> 2021-07-03T11:09:39.004.006.387 (XO: 9.004007802)
   Max TM value:232 at an offset of 39.400 usec -> 2021-07-03T11:09:39.004.009.125 (XO: 9.004010539)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 39.263 usec -> 2021-07-03T11:09:39.004.008.987 (XO: 9.004010402)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 36.500 usec -> 2021-07-03T11:09:39.004.006.225 (XO: 9.004007639)
