-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 29 21:33:04 2021 UTC
-- File 20210703_110742_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.738728 to 20.186949
-each separately from 3s to 20s and from 186660us to 739017us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.447643 in 1 packets: 18
MIU packet time from 2021-07-03T11:07:44.738.728.000 to 2021-07-03T11:08:01.186.949.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.447643 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9261988 to 21.9261378
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999390 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210703_110742_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.6649267 to 5.6652157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4475830  -- OK for SL --
   MIU time from 2021-07-03T11:07:44.738.728.000 to 2021-07-03T11:07:44.739.017.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.447643

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.1127987 to 22.1130877
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-03T11:08:01.186.660.000 to 2021-07-03T11:08:01.186.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.664926700 corrected by 40bit 10MHz counter of 4.926198800 obtained at 1Hz counter of 3. Result: 3.738730559
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-03T11:07:44.738.587.725
   Max 1st sample MIU time among all SL packets: 2021-07-03T11:08:01.186.808.725
   Min 1st sample XO time among all SL packets: 3.738590284
   Max 1st sample XO time among all SL packets: 20.186810298
   TM values from -248 to 219 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-03T11:07:44.738.587.725 (XO: 3.738590284 - 3.738896285, t=35.6 degC)  Rank: 00085
   TM values: Max =   219  Min =  -248  Ave =    -4.39  Std =    19.44
   Min TM value:-248 at an offset of 129.113 usec -> 2021-07-03T11:07:44.738.716.837 (XO: 3.738719397)
   Max TM value:219 at an offset of 131.825 usec -> 2021-07-03T11:07:44.738.719.550 (XO: 3.738722109)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 131.700 usec -> 2021-07-03T11:07:44.738.719.425 (XO: 3.738721984)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 128.950 usec -> 2021-07-03T11:07:44.738.716.675 (XO: 3.738719234)

 - Block number: 2 starting at 2021-07-03T11:08:01.186.519.725 (XO: 20.186521296 - 20.186827298, t=35.6 degC)  Rank: 00079
   TM values: Max =   199  Min =  -241  Ave =    -4.38  Std =    17.87
   Min TM value:-241 at an offset of 131.150 usec -> 2021-07-03T11:08:01.186.650.875 (XO: 20.186652447)
   Max TM value:199 at an offset of 133.838 usec -> 2021-07-03T11:08:01.186.653.562 (XO: 20.186655134)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 133.675 usec -> 2021-07-03T11:08:01.186.653.400 (XO: 20.186654972)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 130.963 usec -> 2021-07-03T11:08:01.186.650.687 (XO: 20.186652259)
