-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 29 20:26:04 2021 UTC
-- File 20210703_093132_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.789538 to 91.636171
-each separately from 1s to 91s and from 107936us to 789827us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.948897 in 1 packets: 18
      20.368923 in 1 packets: 36
      7.679401 in 1 packets: 54
      13.409090 in 1 packets: 72
      19.369124 in 1 packets: 90
      11.069175 in 1 packets: 108
MIU packet time from 2021-07-03T09:31:32.789.538.000 to 2021-07-03T09:33:02.636.171.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.948897 in 1 packets: 18
      20.368923 in 1 packets: 36
      7.679401 in 1 packets: 54
      13.409090 in 1 packets: 72
      19.369124 in 1 packets: 90
      11.069175 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9306797 to 92.9303518
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999345 in 1 packets: 18
      20.9999231 in 1 packets: 36
      6.9999744 in 1 packets: 54
      13.9999487 in 1 packets: 72
      18.9999312 in 1 packets: 90
      10.9999602 in 1 packets: 108
Temperature FPGA: 
      06 -> 47.2 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C0 -> 35.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210703_093132_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7202181 to 3.7205071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.9488310  -- OK for SL --
   MIU time from 2021-07-03T09:31:32.789.538.000 to 2021-07-03T09:31:32.789.827.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.948897

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6693381 to 21.6696271
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3688470  -- OK for SL --
   MIU time from 2021-07-03T09:31:50.738.724.000 to 2021-07-03T09:31:50.739.013.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.368923

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.0384741 to 42.0387631
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6793750  -- OK for SL --
   MIU time from 2021-07-03T09:32:11.107.936.000 to 2021-07-03T09:32:11.108.225.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.679401

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.7181381 to 49.7184271
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4090390  -- OK for SL --
   MIU time from 2021-07-03T09:32:18.787.626.000 to 2021-07-03T09:32:18.787.915.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.409090

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.1274661 to 63.1277551
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3690550  -- OK for SL --
   MIU time from 2021-07-03T09:32:32.197.005.000 to 2021-07-03T09:32:32.197.294.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.369124

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.4968101 to 82.4970991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0691350  -- OK for SL --
   MIU time from 2021-07-03T09:32:51.566.418.000 to 2021-07-03T09:32:51.566.707.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.069175

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 93.5662341 to 93.5665231
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-03T09:33:02.635.882.000 to 2021-07-03T09:33:02.636.171.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.720218100 corrected by 40bit 10MHz counter of 2.930679700 obtained at 1Hz counter of 1. Result: 1.789541242
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-03T09:31:32.789.397.725
   Max 1st sample MIU time among all SL packets: 2021-07-03T09:33:02.636.030.725
   Min 1st sample XO time among all SL packets: 1.789400967
   Max 1st sample XO time among all SL packets: 91.636033315
   TM values from -277 to 264 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-03T09:31:32.789.397.725 (XO: 1.789400967 - 1.789706968, t=35.6 degC)  Rank: 00078
   TM values: Max =   177  Min =  -245  Ave =    -4.56  Std =    18.97
   Min TM value:-245 at an offset of 108.313 usec -> 2021-07-03T09:31:32.789.506.037 (XO: 1.789509280)
   Max TM value:177 at an offset of 107.988 usec -> 2021-07-03T09:31:32.789.505.712 (XO: 1.789508955)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 110.788 usec -> 2021-07-03T09:31:32.789.508.512 (XO: 1.789511755)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 108.163 usec -> 2021-07-03T09:31:32.789.505.887 (XO: 1.789509130)

 - Block number: 2 starting at 2021-07-03T09:31:50.738.583.725 (XO: 19.738586284 - 19.738892285, t=35.6 degC)  Rank: 00072
   TM values: Max =   264  Min =  -258  Ave =    -4.39  Std =    24.91
   Min TM value:-258 at an offset of 121.075 usec -> 2021-07-03T09:31:50.738.704.800 (XO: 19.738707359)
   Max TM value:264 at an offset of 120.788 usec -> 2021-07-03T09:31:50.738.704.512 (XO: 19.738707072)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 118.763 usec -> 2021-07-03T09:31:50.738.702.487 (XO: 19.738705047)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 120.900 usec -> 2021-07-03T09:31:50.738.704.625 (XO: 19.738707184)

 - Block number: 3 starting at 2021-07-03T09:32:11.107.795.725 (XO: 40.107796913 - 40.108102914, t=35.6 degC)  Rank: 00083
   TM values: Max =   212  Min =  -260  Ave =    -4.33  Std =    20.62
   Min TM value:-260 at an offset of 119.625 usec -> 2021-07-03T09:32:11.107.915.350 (XO: 40.107916538)
   Max TM value:212 at an offset of 122.288 usec -> 2021-07-03T09:32:11.107.918.012 (XO: 40.107919201)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 122.113 usec -> 2021-07-03T09:32:11.107.917.837 (XO: 40.107919026)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 119.388 usec -> 2021-07-03T09:32:11.107.915.112 (XO: 40.107916301)

 - Block number: 4 starting at 2021-07-03T09:32:18.787.485.725 (XO: 47.787489039 - 47.787795040, t=35.6 degC)  Rank: 00076
   TM values: Max =   245  Min =  -277  Ave =    -4.47  Std =    26.35
   Min TM value:-277 at an offset of 107.350 usec -> 2021-07-03T09:32:18.787.593.075 (XO: 47.787596389)
   Max TM value:245 at an offset of 107.050 usec -> 2021-07-03T09:32:18.787.592.775 (XO: 47.787596089)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 109.775 usec -> 2021-07-03T09:32:18.787.595.500 (XO: 47.787598814)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 107.175 usec -> 2021-07-03T09:32:18.787.592.900 (XO: 47.787596214)

 - Block number: 5 starting at 2021-07-03T09:32:32.196.864.725 (XO: 61.196866153 - 61.197172155, t=35.6 degC)  Rank: 00079
   TM values: Max =   221  Min =  -232  Ave =    -4.43  Std =    22.34
   Min TM value:-232 at an offset of 178.288 usec -> 2021-07-03T09:32:32.197.043.013 (XO: 61.197044442)
   Max TM value:221 at an offset of 120.013 usec -> 2021-07-03T09:32:32.196.984.737 (XO: 61.196986166)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 119.850 usec -> 2021-07-03T09:32:32.196.984.575 (XO: 61.196986004)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 178.126 usec -> 2021-07-03T09:32:32.197.042.850 (XO: 61.197044279)

 - Block number: 6 starting at 2021-07-03T09:32:51.566.277.725 (XO: 80.566280320 - 80.566586321, t=35.6 degC)  Rank: 00080
   TM values: Max =   209  Min =  -246  Ave =    -4.44  Std =    19.36
   Min TM value:-246 at an offset of 119.850 usec -> 2021-07-03T09:32:51.566.397.575 (XO: 80.566400171)
   Max TM value:209 at an offset of 119.513 usec -> 2021-07-03T09:32:51.566.397.237 (XO: 80.566399833)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 122.213 usec -> 2021-07-03T09:32:51.566.399.937 (XO: 80.566402533)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 119.625 usec -> 2021-07-03T09:32:51.566.397.350 (XO: 80.566399946)

 - Block number: 7 starting at 2021-07-03T09:33:02.635.741.725 (XO: 91.635744314 - 91.636050315, t=35.6 degC)  Rank: 00085
   TM values: Max =   207  Min =  -229  Ave =    -4.51  Std =    19.82
   Min TM value:-229 at an offset of 117.288 usec -> 2021-07-03T09:33:02.635.859.012 (XO: 91.635861602)
   Max TM value:207 at an offset of 120.050 usec -> 2021-07-03T09:33:02.635.861.775 (XO: 91.635864364)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 119.938 usec -> 2021-07-03T09:33:02.635.861.662 (XO: 91.635864252)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 104.800 usec -> 2021-07-03T09:33:02.635.846.525 (XO: 91.635849114)
