-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 10:20:23 2023 UTC
-- File 20210702_155001_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.946450 to 69.376761
-each separately from 9s to 69s and from 276401us to 946739us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.329662 in 1 packets: 18
      17.089644 in 1 packets: 36
      19.379684 in 1 packets: 54
      14.629876 in 1 packets: 72
MIU packet time from 2021-07-02T15:50:09.946.450.000 to 2021-07-02T15:51:09.376.761.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.329662 in 1 packets: 18
      17.089644 in 1 packets: 36
      19.379684 in 1 packets: 54
      14.629876 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8538369 to 70.8536222
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999677 in 1 packets: 18
      16.9999392 in 1 packets: 36
      18.9999320 in 1 packets: 54
      14.9999464 in 1 packets: 72
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      3 in 1 packets: 1
      4 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      7 in 1 packets: 19
      8 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      14 in 1 packets: 37
      15 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      18 in 1 packets: 55
      19 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      20 in 1 packets: 73
      21 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      3 in 1 packets: 1
      4 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      7 in 1 packets: 19
      8 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      14 in 1 packets: 37
      15 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      18 in 1 packets: 55
      19 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      20 in 1 packets: 73
      21 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210702_155001_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8002866 to 11.8005756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3296310  -- OK for SL --
   MIU time from 2021-07-02T15:50:09.946.450.000 to 2021-07-02T15:50:09.946.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.329662

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.1302066 to 20.1304956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0895830  -- OK for SL --
   MIU time from 2021-07-02T15:50:18.276.401.000 to 2021-07-02T15:50:18.276.690.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.089644

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2200786 to 37.2203676
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3796150  -- OK for SL --
   MIU time from 2021-07-02T15:50:35.366.334.000 to 2021-07-02T15:50:35.366.623.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.379684

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.5999826 to 56.6002716
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6298230  -- OK for SL --
   MIU time from 2021-07-02T15:50:54.746.307.000 to 2021-07-02T15:50:54.746.596.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.629876

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.2300946 to 71.2303836
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-02T15:51:09.376.472.000 to 2021-07-02T15:51:09.376.761.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.800286600 corrected by 40bit 10MHz counter of 10.853836900 obtained at 1Hz counter of 9. Result: 9.946453107
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-02T15:50:09.946.309.725
   Max 1st sample MIU time among all SL packets: 2021-07-02T15:51:09.376.620.725
   Min 1st sample XO time among all SL packets: 9.946312832
   Max 1st sample XO time among all SL packets: 69.376622481
   TM values from -275 to 259 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-02T15:50:09.946.309.725 (XO: 9.946312832 - 9.946618833, t=35.6 degC)  Rank: 00088
   TM values: Max =   259  Min =  -233  Ave =    -4.34  Std =    23.54
   Min TM value:-233 at an offset of 118.813 usec -> 2021-07-02T15:50:09.946.428.537 (XO: 9.946431645)
   Max TM value:259 at an offset of 119.063 usec -> 2021-07-02T15:50:09.946.428.787 (XO: 9.946431895)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 118.938 usec -> 2021-07-02T15:50:09.946.428.662 (XO: 9.946431770)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 116.225 usec -> 2021-07-02T15:50:09.946.425.950 (XO: 9.946429057)

 - Block number: 2 starting at 2021-07-02T15:50:18.276.260.725 (XO: 18.276262692 - 18.276568693, t=35.6 degC)  Rank: 00084
   TM values: Max =   215  Min =  -251  Ave =    -4.44  Std =    20.07
   Min TM value:-251 at an offset of 128.688 usec -> 2021-07-02T15:50:18.276.389.412 (XO: 18.276391380)
   Max TM value:215 at an offset of 131.313 usec -> 2021-07-02T15:50:18.276.392.037 (XO: 18.276394005)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 131.150 usec -> 2021-07-02T15:50:18.276.391.875 (XO: 18.276393842)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 128.488 usec -> 2021-07-02T15:50:18.276.389.212 (XO: 18.276391180)

 - Block number: 3 starting at 2021-07-02T15:50:35.366.193.725 (XO: 35.366195843 - 35.366501844, t=35.6 degC)  Rank: 00087
   TM values: Max =   207  Min =  -275  Ave =    -4.43  Std =    22.01
   Min TM value:-275 at an offset of 132.375 usec -> 2021-07-02T15:50:35.366.326.100 (XO: 35.366328219)
   Max TM value:207 at an offset of 135.038 usec -> 2021-07-02T15:50:35.366.328.762 (XO: 35.366330881)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 129.925 usec -> 2021-07-02T15:50:35.366.323.650 (XO: 35.366325769)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 132.250 usec -> 2021-07-02T15:50:35.366.325.975 (XO: 35.366328094)

 - Block number: 4 starting at 2021-07-02T15:50:54.746.166.725 (XO: 54.746169137 - 54.746475138, t=35.6 degC)  Rank: 00084
   TM values: Max =   214  Min =  -267  Ave =    -4.34  Std =    29.93
   Min TM value:-267 at an offset of 118.488 usec -> 2021-07-02T15:50:54.746.285.212 (XO: 54.746287625)
   Max TM value:214 at an offset of 118.213 usec -> 2021-07-02T15:50:54.746.284.937 (XO: 54.746287350)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 160.801 usec -> 2021-07-02T15:50:54.746.327.525 (XO: 54.746329937)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 118.350 usec -> 2021-07-02T15:50:54.746.285.075 (XO: 54.746287487)

 - Block number: 5 starting at 2021-07-02T15:51:09.376.331.725 (XO: 69.376333480 - 69.376639481, t=35.6 degC)  Rank: 00087
   TM values: Max =   233  Min =  -250  Ave =    -4.29  Std =    24.26
   Min TM value:-250 at an offset of 128.125 usec -> 2021-07-02T15:51:09.376.459.850 (XO: 69.376461605)
   Max TM value:233 at an offset of 130.750 usec -> 2021-07-02T15:51:09.376.462.475 (XO: 69.376464230)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 130.663 usec -> 2021-07-02T15:51:09.376.462.387 (XO: 69.376464143)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 127.938 usec -> 2021-07-02T15:51:09.376.459.662 (XO: 69.376461418)
