-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 29 11:14:41 2021 UTC
-- File 20210702_113326_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.981197 to 17.680666
-each separately from 9s to 17s and from 680377us to 981486us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.698891 in 1 packets: 18
MIU packet time from 2021-07-02T11:33:34.981.197.000 to 2021-07-02T11:33:42.680.666.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.698891 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8829389 to 18.8829103
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999714 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210702_113326_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8641360 to 11.8644250
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6988630  -- OK for SL --
   MIU time from 2021-07-02T11:33:34.981.197.000 to 2021-07-02T11:33:34.981.486.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.698891

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.5632880 to 19.5635770
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-02T11:33:42.680.377.000 to 2021-07-02T11:33:42.680.666.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.864136000 corrected by 40bit 10MHz counter of 10.882938900 obtained at 1Hz counter of 9. Result: 9.981200632
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-02T11:33:34.981.056.725
   Max 1st sample MIU time among all SL packets: 2021-07-02T11:33:42.680.525.725
   Min 1st sample XO time among all SL packets: 9.981060357
   Max 1st sample XO time among all SL packets: 17.680528875
   TM values from -252 to 231 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-02T11:33:34.981.056.725 (XO: 9.981060357 - 9.981366358, t=35.6 degC)  Rank: 00080
   TM values: Max =   231  Min =  -231  Ave =    -4.48  Std =    22.06
   Min TM value:-231 at an offset of 120.125 usec -> 2021-07-02T11:33:34.981.176.850 (XO: 9.981180482)
   Max TM value:231 at an offset of 122.788 usec -> 2021-07-02T11:33:34.981.179.512 (XO: 9.981183145)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 122.650 usec -> 2021-07-02T11:33:34.981.179.375 (XO: 9.981183007)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 119.938 usec -> 2021-07-02T11:33:34.981.176.662 (XO: 9.981180295)

 - Block number: 2 starting at 2021-07-02T11:33:42.680.236.725 (XO: 17.680239874 - 17.680545875, t=35.6 degC)  Rank: 00088
   TM values: Max =   228  Min =  -252  Ave =    -4.31  Std =    21.94
   Min TM value:-252 at an offset of 106.113 usec -> 2021-07-02T11:33:42.680.342.837 (XO: 17.680345987)
   Max TM value:228 at an offset of 121.363 usec -> 2021-07-02T11:33:42.680.358.087 (XO: 17.680361237)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 121.213 usec -> 2021-07-02T11:33:42.680.357.937 (XO: 17.680361087)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 118.538 usec -> 2021-07-02T11:33:42.680.355.262 (XO: 17.680358412)
