-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 13:57:00 2021 UTC
-- File 20210702_021802_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.360055 to 38.670435
-each separately from 8s to 38s and from 360055us to 670435us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.059738 in 1 packets: 18
      9.249775 in 1 packets: 36
MIU packet time from 2021-07-02T02:18:09.360.055.000 to 2021-07-02T02:18:39.670.435.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.059738 in 1 packets: 18
      9.249775 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8591301 to 38.8590233
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999253 in 1 packets: 18
      8.9999679 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210702_021802_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2191857 to 9.2194747
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0596630  -- OK for SL --
   MIU time from 2021-07-02T02:18:09.360.055.000 to 2021-07-02T02:18:09.360.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.059738

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.2791377 to 30.2794267
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2497430  -- OK for SL --
   MIU time from 2021-07-02T02:18:30.420.082.000 to 2021-07-02T02:18:30.420.371.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.249775

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.5291697 to 39.5294587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-02T02:18:39.670.146.000 to 2021-07-02T02:18:39.670.435.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 45 XO calibration points at 1Hz counter from 0 to 44
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.219185700 corrected by 40bit 10MHz counter of 8.859130100 obtained at 1Hz counter of 8. Result: 8.360056860
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-02T02:18:09.359.914.725
   Max 1st sample MIU time among all SL packets: 2021-07-02T02:18:39.670.294.725
   Min 1st sample XO time among all SL packets: 8.359916585
   Max 1st sample XO time among all SL packets: 38.670297538
   TM values from -233 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-02T02:18:09.359.914.725 (XO: 8.359916585 - 8.360222586, t=35.6 degC)  Rank: 00080
   TM values: Max =   235  Min =  -233  Ave =    -4.43  Std =    22.00
   Min TM value:-233 at an offset of 128.325 usec -> 2021-07-02T02:18:09.360.043.050 (XO: 8.360044910)
   Max TM value:235 at an offset of 131.063 usec -> 2021-07-02T02:18:09.360.045.787 (XO: 8.360047648)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 130.950 usec -> 2021-07-02T02:18:09.360.045.675 (XO: 8.360047535)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 128.200 usec -> 2021-07-02T02:18:09.360.042.925 (XO: 8.360044785)

 - Block number: 2 starting at 2021-07-02T02:18:30.419.941.725 (XO: 29.419943537 - 29.420249538, t=35.6 degC)  Rank: 00078
   TM values: Max =   230  Min =  -229  Ave =    -4.52  Std =    19.83
   Min TM value:-229 at an offset of 129.763 usec -> 2021-07-02T02:18:30.420.071.487 (XO: 29.420073300)
   Max TM value:230 at an offset of 132.475 usec -> 2021-07-02T02:18:30.420.074.200 (XO: 29.420076012)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 132.363 usec -> 2021-07-02T02:18:30.420.074.087 (XO: 29.420075900)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 129.575 usec -> 2021-07-02T02:18:30.420.071.300 (XO: 29.420073112)

 - Block number: 3 starting at 2021-07-02T02:18:39.670.005.725 (XO: 38.670008537 - 38.670314538, t=35.6 degC)  Rank: 00081
   TM values: Max =   205  Min =  -215  Ave =    -4.45  Std =    19.01
   Min TM value:-215 at an offset of 112.425 usec -> 2021-07-02T02:18:39.670.118.150 (XO: 38.670120962)
   Max TM value:205 at an offset of 127.625 usec -> 2021-07-02T02:18:39.670.133.350 (XO: 38.670136162)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 127.450 usec -> 2021-07-02T02:18:39.670.133.175 (XO: 38.670135987)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 112.300 usec -> 2021-07-02T02:18:39.670.118.025 (XO: 38.670120837)
