-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Aug  1 03:59:56 2021 UTC
-- File 20210630_200647_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.049606 to 42.462054
-each separately from 2s to 42s and from 397us to 462054us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.950502 in 1 packets: 18
      20.461079 in 1 packets: 36
MIU packet time from 2021-06-30T20:06:48.049.606.000 to 2021-06-30T20:07:28.462.054.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.950502 in 1 packets: 18
      20.461079 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8260549 to 43.8259126
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999289 in 1 packets: 18
      19.9999288 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210630_200647_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8756622 to 3.8759512
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9504310  -- OK for SL --
   MIU time from 2021-06-30T20:06:48.049.606.000 to 2021-06-30T20:06:48.049.895.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.950502

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8263822 to 23.8266712
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4610070  -- OK for SL --
   MIU time from 2021-06-30T20:07:08.000.397.000 to 2021-06-30T20:07:08.000.686.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.461079

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.2876782 to 44.2879672
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-30T20:07:28.461.765.000 to 2021-06-30T20:07:28.462.054.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.875662200 corrected by 40bit 10MHz counter of 3.826054900 obtained at 1Hz counter of 2. Result: 2.049607479
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-06-30T20:06:48.049.465.725
   Max 1st sample MIU time among all SL packets: 2021-06-30T20:07:28.461.913.725
   Min 1st sample XO time among all SL packets: 2.049467203
   Max 1st sample XO time among all SL packets: 42.461915942
   TM values from -279 to 260 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-30T20:06:48.049.465.725 (XO: 2.049467203 - 2.049773204, t=35.6 degC)  Rank: 00083
   TM values: Max =   260  Min =  -271  Ave =    -4.39  Std =    25.76
   Min TM value:-271 at an offset of 108.900 usec -> 2021-06-30T20:06:48.049.574.625 (XO: 2.049576103)
   Max TM value:260 at an offset of 111.538 usec -> 2021-06-30T20:06:48.049.577.262 (XO: 2.049578741)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 111.488 usec -> 2021-06-30T20:06:48.049.577.212 (XO: 2.049578691)
   Min dTM/dt [1/us]:-2.72e+03 at an offset of 108.763 usec -> 2021-06-30T20:06:48.049.574.487 (XO: 2.049575966)

 - Block number: 2 starting at 2021-06-30T20:07:08.000.256.725 (XO: 22.000258126 - 22.000564127, t=35.6 degC)  Rank: 00079
   TM values: Max =   210  Min =  -272  Ave =    -4.41  Std =    22.77
   Min TM value:-272 at an offset of 124.238 usec -> 2021-06-30T20:07:08.000.380.962 (XO: 22.000382364)
   Max TM value:210 at an offset of 126.788 usec -> 2021-06-30T20:07:08.000.383.512 (XO: 22.000384914)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 126.600 usec -> 2021-06-30T20:07:08.000.383.325 (XO: 22.000384726)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 124.075 usec -> 2021-06-30T20:07:08.000.380.800 (XO: 22.000382201)

 - Block number: 3 starting at 2021-06-30T20:07:28.461.624.725 (XO: 42.461626941 - 42.461932942, t=35.6 degC)  Rank: 00084
   TM values: Max =   249  Min =  -279  Ave =    -4.33  Std =    27.75
   Min TM value:-279 at an offset of 126.138 usec -> 2021-06-30T20:07:28.461.750.862 (XO: 42.461753079)
   Max TM value:249 at an offset of 123.575 usec -> 2021-06-30T20:07:28.461.748.300 (XO: 42.461750516)
   Max dTM/dt [1/us]:2.59e+03 at an offset of 126.263 usec -> 2021-06-30T20:07:28.461.750.987 (XO: 42.461753204)
   Min dTM/dt [1/us]:-2.99e+03 at an offset of 123.713 usec -> 2021-06-30T20:07:28.461.748.437 (XO: 42.461750654)
