-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul 24 03:11:05 2021 UTC
-- File 20210629_165239_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.077986 to 20.432943
-each separately from 8s to 20s and from 77986us to 432943us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.354379 in 1 packets: 18
MIU packet time from 2021-06-29T16:52:46.077.986.000 to 2021-06-29T16:52:58.432.943.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.354379 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4497617 to 21.4497185
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999568 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210629_165239_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5277489 to 9.5280379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3543350  -- OK for SL --
   MIU time from 2021-06-29T16:52:46.077.986.000 to 2021-06-29T16:52:46.078.275.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.354379

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.8823729 to 21.8826619
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-29T16:52:58.432.654.000 to 2021-06-29T16:52:58.432.943.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.527748900 corrected by 40bit 10MHz counter of 9.449761700 obtained at 1Hz counter of 8. Result: 8.077987481
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-29T16:52:46.077.845.725
   Max 1st sample MIU time among all SL packets: 2021-06-29T16:52:58.432.802.725
   Min 1st sample XO time among all SL packets: 8.077847205
   Max 1st sample XO time among all SL packets: 20.432804683
   TM values from -334 to 271 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-29T16:52:46.077.845.725 (XO: 8.077847205 - 8.078153206, t=35.6 degC)  Rank: 00086
   TM values: Max =   140  Min =  -158  Ave =    -4.48  Std =    54.04
   Min TM value:-158 at an offset of 96.463 usec -> 2021-06-29T16:52:46.077.942.187 (XO: 8.077943668)
   Max TM value:140 at an offset of 91.325 usec -> 2021-06-29T16:52:46.077.937.050 (XO: 8.077938531)
   Max dTM/dt [1/us]:1.20e+03 at an offset of 305.989 usec -> 2021-06-29T16:52:46.078.151.713 (XO: 8.078153194)
   Min dTM/dt [1/us]:-1.18e+03 at an offset of 197.426 usec -> 2021-06-29T16:52:46.078.043.150 (XO: 8.078044631)

 - Block number: 2 starting at 2021-06-29T16:52:58.432.513.725 (XO: 20.432515682 - 20.432821683, t=35.6 degC)  Rank: 00078
   TM values: Max =   271  Min =  -334  Ave =    -4.48  Std =    34.85
   Min TM value:-334 at an offset of 112.963 usec -> 2021-06-29T16:52:58.432.626.687 (XO: 20.432628645)
   Max TM value:271 at an offset of 110.625 usec -> 2021-06-29T16:52:58.432.624.350 (XO: 20.432626307)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 110.513 usec -> 2021-06-29T16:52:58.432.624.237 (XO: 20.432626195)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 112.850 usec -> 2021-06-29T16:52:58.432.626.575 (XO: 20.432628532)
