-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 08:46:51 2023 UTC
-- File 20210629_085438_1229.dat, 7512  packets found.
APID:
      1229 (EW) in 7512 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 7512 packets (all packets)
Total number of packets: 
      36720 in 7512 packets (all packets)
Packet No from 1 to 7512
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 213.343904 to 213.343904
-each separately from 213s to 213s and from 343904us to 343904us.
   Differences in neigboring packets: 
      0.000000 in 7511 packets (all packets)
MIU packet time from 2021-06-29T08:58:11.343.904.000 to 2021-06-29T08:58:11.343.904.000
   Differences in neigboring packets: 
      0.000000 in 7511 packets (all packets)
MIU orbit number: 
      60000 in 7512 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 7512 packets (all packets)
MIU orbit information / number of events: 
      0 in 7512 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 7512 packets (all packets)
Instrument anomalies: 
      0 (OK) in 7512 packets (all packets)
FPGA configuration version: 
      211 in 7512 packets (all packets)
10 MHz counter at the last 1Hz pulse from 214.8085311 to 214.8085311
   Differences in neigboring packets: 
      0.0000000 in 7511 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 7512 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 7512 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 7512 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 7512 packets (all packets)
Number of external events: 
      3 in 7512 packets (all packets)
Number of generated alerts: 
      3 in 7512 packets (all packets)
Survey memory error: 
      0 in 7512 packets (all packets)
Event memory error: 
      0 in 7512 packets (all packets)
Configuration error: 
      0 in 7512 packets (all packets)
SDRAM CRC error: 
      0 in 7512 packets (all packets)
Radio gain: 
      0 in 7512 packets (all packets)
Radio bandwidth: 
      0 in 7512 packets (all packets)
Digital gain: 
      0 in 7512 packets (all packets)
Analog gain: 
      0 in 7512 packets (all packets)

EVENT DATA READER
-- File 20210629_085438_1229.dat, 36 EW pages found. Extra packets: 168.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 3 to 3
      3 in 7344 packets (all packets)
   LS 32 bits of C10MHz from descriptor (period 430s)  from 215.0969483 to 215.2217793
      Difference between the neighboring packets: 
      0.0000170 in 7343 packets (all packets)
   Memory page number from 2 to 37
      Difference between the neighboring packets: 
      0 in 7308 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 36
      Difference between the neighboring packets: 
      0 in 7308 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   MIU event flag 
      1 in 7344 packets (all packets)
   Time triggered event flag 
      0 in 7344 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2021-06-29T08:58:11.288.415.626
   Min EW header abs. time found among all events            : 2021-06-29T08:58:11.343.904.000
   Max EW event (1st sample) abs. time found among all events: 2021-06-29T08:58:11.288.415.626
   Max EW header abs. time found among all events            : 2021-06-29T08:58:11.343.904.000
   TM values from -264 to 220 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 2 total number of packets: 0
   No packets in this event  -- NOT OK -- 

 Event no. 3 total number of packets: 7344
    Page no. 1:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 36  ordered packets: 7344
   TM values from -264 to 220 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=213 C1MHz=0.343904
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 7343 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 215.0969483
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2021-06-29T08:58:11.343.904.000
   Event 1st sample abs. time : 2021-06-29T08:58:11.288.415.626)
   --- Rel. time correction for 1 points based on 209 XO calibration points at 1Hz counter from 0 to 208
   --- First point: 1Hz counter = 213, 32bit 10MHz counter = 215.096948300 corrected by 40bit 10MHz counter of 209.808549200 obtained at 1Hz counter of 208. Result: 213.288418138
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):             213.288417963
   Event data descriptor XO time (reconstructed seconds from XP start)    :             213.288418138
   Sampling frequency correction factor for this event:     0.999996400
--
Extremes in separate events:

 - Event number: 1 --EMPTY-- 

 - Event number: 2 --EMPTY-- 

 - Event number: 3 from 2021-06-29T08:58:11.288.415.626 to 2021-06-29T08:58:11.413.264.075 (XO: 213.288417963 - 213.413266413)
   Min TM value:-264 at an offset of 53279.092 usec -> 2021-06-29T08:58:11.341.694.717 (XO: 213.341697055)
   Max TM value:220 at an offset of 53276.554 usec -> 2021-06-29T08:58:11.341.692.180 (XO: 213.341694518)
   Max dTM/dt [1/us]:2.32e+03 at an offset of 53279.204 usec -> 2021-06-29T08:58:11.341.694.830 (XO: 213.341697168)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 53276.667 usec -> 2021-06-29T08:58:11.341.692.292 (XO: 213.341694630)
   Med: -4.00000   Ave: -4.4754875   Std: 9.9460808
