-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul 23 15:58:57 2021 UTC
-- File 20210629_052116_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.504176 to 87.625421
-each separately from 13s to 87s and from 14368us to 964719us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.509903 in 1 packets: 18
      8.949773 in 1 packets: 36
      20.190074 in 1 packets: 54
      14.989926 in 1 packets: 72
      21.479835 in 1 packets: 90
MIU packet time from 2021-06-29T05:21:29.504.176.000 to 2021-06-29T05:22:43.625.421.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.509903 in 1 packets: 18
      8.949773 in 1 packets: 36
      20.190074 in 1 packets: 54
      14.989926 in 1 packets: 72
      21.479835 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9179727 to 88.9177041
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999673 in 1 packets: 18
      7.9999710 in 1 packets: 36
      20.9999238 in 1 packets: 54
      14.9999456 in 1 packets: 72
      20.9999237 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C0 -> 35.6 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 107 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 107 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210629_052116_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.4221489 to 15.4224379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.5098710  -- OK for SL --
   MIU time from 2021-06-29T05:21:29.504.176.000 to 2021-06-29T05:21:29.504.465.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.509903

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.9323089 to 23.9325979
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9497430  -- OK for SL --
   MIU time from 2021-06-29T05:21:38.014.368.000 to 2021-06-29T05:21:38.014.657.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.949773

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.8823409 to 32.8826299
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1899990  -- OK for SL --
   MIU time from 2021-06-29T05:21:46.964.430.000 to 2021-06-29T05:21:46.964.719.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.190074

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.0726289 to 53.0729179
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9898710  -- OK for SL --
   MIU time from 2021-06-29T05:22:07.154.793.000 to 2021-06-29T05:22:07.155.082.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.989926

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.0627889 to 68.0630779
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4797590  -- OK for SL --
   MIU time from 2021-06-29T05:22:22.145.008.000 to 2021-06-29T05:22:22.145.297.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.479835

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 89.5428369 to 89.5431259
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-29T05:22:43.625.132.000 to 2021-06-29T05:22:43.625.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.422148900 corrected by 40bit 10MHz counter of 14.917972700 obtained at 1Hz counter of 13. Result: 13.504178015
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-29T05:21:29.504.035.725
   Max 1st sample MIU time among all SL packets: 2021-06-29T05:22:43.625.280.725
   Min 1st sample XO time among all SL packets: 13.504037740
   Max 1st sample XO time among all SL packets: 87.625283776
   TM values from -245 to 234 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-29T05:21:29.504.035.725 (XO: 13.504037740 - 13.504343741, t=35.6 degC)  Rank: 00086
   TM values: Max =   199  Min =  -236  Ave =    -4.42  Std =    20.83
   Min TM value:-236 at an offset of 107.650 usec -> 2021-06-29T05:21:29.504.143.375 (XO: 13.504145390)
   Max TM value:199 at an offset of 94.825 usec -> 2021-06-29T05:21:29.504.130.550 (XO: 13.504132565)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 110.213 usec -> 2021-06-29T05:21:29.504.145.937 (XO: 13.504147952)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 94.950 usec -> 2021-06-29T05:21:29.504.130.675 (XO: 13.504132690)

 - Block number: 2 starting at 2021-06-29T05:21:38.014.227.725 (XO: 22.014228676 - 22.014534677, t=35.6 degC)  Rank: 00077
   TM values: Max =   223  Min =  -240  Ave =    -4.40  Std =    21.00
   Min TM value:-240 at an offset of 96.713 usec -> 2021-06-29T05:21:38.014.324.437 (XO: 22.014325389)
   Max TM value:223 at an offset of 94.150 usec -> 2021-06-29T05:21:38.014.321.875 (XO: 22.014322827)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 96.850 usec -> 2021-06-29T05:21:38.014.324.575 (XO: 22.014325527)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 94.238 usec -> 2021-06-29T05:21:38.014.321.962 (XO: 22.014322914)

 - Block number: 3 starting at 2021-06-29T05:21:46.964.289.725 (XO: 30.964293193 - 30.964599194, t=35.6 degC)  Rank: 00079
   TM values: Max =   177  Min =  -235  Ave =    -4.51  Std =    25.11
   Min TM value:-235 at an offset of 93.050 usec -> 2021-06-29T05:21:46.964.382.775 (XO: 30.964386243)
   Max TM value:177 at an offset of 92.688 usec -> 2021-06-29T05:21:46.964.382.412 (XO: 30.964385881)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 108.363 usec -> 2021-06-29T05:21:46.964.398.087 (XO: 30.964401556)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 105.688 usec -> 2021-06-29T05:21:46.964.395.412 (XO: 30.964398881)

 - Block number: 4 starting at 2021-06-29T05:22:07.154.652.725 (XO: 51.154654382 - 51.154960383, t=35.6 degC)  Rank: 00076
   TM values: Max =   190  Min =  -233  Ave =    -4.40  Std =    17.83
   Min TM value:-233 at an offset of 120.400 usec -> 2021-06-29T05:22:07.154.773.125 (XO: 51.154774782)
   Max TM value:190 at an offset of 123.075 usec -> 2021-06-29T05:22:07.154.775.800 (XO: 51.154777457)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 122.900 usec -> 2021-06-29T05:22:07.154.775.625 (XO: 51.154777282)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 120.238 usec -> 2021-06-29T05:22:07.154.772.962 (XO: 51.154774620)

 - Block number: 5 starting at 2021-06-29T05:22:22.144.867.725 (XO: 66.144868761 - 66.145174762, t=35.6 degC)  Rank: 00077
   TM values: Max =   218  Min =  -245  Ave =    -4.46  Std =    20.85
   Min TM value:-245 at an offset of 123.825 usec -> 2021-06-29T05:22:22.144.991.550 (XO: 66.144992586)
   Max TM value:218 at an offset of 123.538 usec -> 2021-06-29T05:22:22.144.991.262 (XO: 66.144992299)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 126.263 usec -> 2021-06-29T05:22:22.144.993.987 (XO: 66.144995024)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 123.688 usec -> 2021-06-29T05:22:22.144.991.412 (XO: 66.144992449)

 - Block number: 6 starting at 2021-06-29T05:22:43.624.991.725 (XO: 87.624994775 - 87.625300776, t=35.6 degC)  Rank: 00078
   TM values: Max =   234  Min =  -242  Ave =    -4.53  Std =    29.64
   Min TM value:-242 at an offset of 176.276 usec -> 2021-06-29T05:22:43.625.168.000 (XO: 87.625171051)
   Max TM value:234 at an offset of 175.988 usec -> 2021-06-29T05:22:43.625.167.713 (XO: 87.625170763)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 173.951 usec -> 2021-06-29T05:22:43.625.165.675 (XO: 87.625168726)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 176.101 usec -> 2021-06-29T05:22:43.625.167.825 (XO: 87.625170876)
