-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul 23 11:13:59 2021 UTC
-- File 20210628_174712_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.476220 to 38.577344
-each separately from 6s to 38s and from 206597us to 577344us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.730088 in 1 packets: 18
      16.370169 in 1 packets: 36
MIU packet time from 2021-06-28T17:47:18.476.220.000 to 2021-06-28T17:47:50.577.344.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.730088 in 1 packets: 18
      16.370169 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.2930872 to 40.2929715
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999422 in 1 packets: 18
      15.9999421 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210628_174712_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7693074 to 8.7695964
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7300310  -- OK for SL --
   MIU time from 2021-06-28T17:47:18.476.220.000 to 2021-06-28T17:47:18.476.509.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.730088

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4996274 to 24.4999164
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3701110  -- OK for SL --
   MIU time from 2021-06-28T17:47:34.206.597.000 to 2021-06-28T17:47:34.206.886.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.370169

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.8700274 to 40.8703164
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-28T17:47:50.577.055.000 to 2021-06-28T17:47:50.577.344.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.769307400 corrected by 40bit 10MHz counter of 8.293087200 obtained at 1Hz counter of 6. Result: 6.476221914
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-28T17:47:18.476.079.725
   Max 1st sample MIU time among all SL packets: 2021-06-28T17:47:50.577.203.725
   Min 1st sample XO time among all SL packets: 6.476081639
   Max 1st sample XO time among all SL packets: 38.577206703
   TM values from -367 to 277 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-28T17:47:18.476.079.725 (XO: 6.476081639 - 6.476387640, t=35.6 degC)  Rank: 00082
   TM values: Max =   277  Min =  -367  Ave =    -4.49  Std =    45.32
   Min TM value:-367 at an offset of 128.588 usec -> 2021-06-28T17:47:18.476.208.312 (XO: 6.476210227)
   Max TM value:277 at an offset of 128.813 usec -> 2021-06-28T17:47:18.476.208.537 (XO: 6.476210452)
   Max dTM/dt [1/us]:2.90e+03 at an offset of 128.700 usec -> 2021-06-28T17:47:18.476.208.425 (XO: 6.476210339)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 126.100 usec -> 2021-06-28T17:47:18.476.205.825 (XO: 6.476207739)

 - Block number: 2 starting at 2021-06-28T17:47:34.206.456.725 (XO: 22.206458468 - 22.206764469, t=35.6 degC)  Rank: 00071
   TM values: Max =   251  Min =  -287  Ave =    -4.45  Std =    37.02
   Min TM value:-287 at an offset of 53.850 usec -> 2021-06-28T17:47:34.206.510.575 (XO: 22.206512318)
   Max TM value:251 at an offset of 53.563 usec -> 2021-06-28T17:47:34.206.510.287 (XO: 22.206512031)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 114.888 usec -> 2021-06-28T17:47:34.206.571.612 (XO: 22.206573356)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 53.725 usec -> 2021-06-28T17:47:34.206.510.450 (XO: 22.206512193)

 - Block number: 3 starting at 2021-06-28T17:47:50.576.914.725 (XO: 38.576917702 - 38.577223703, t=35.6 degC)  Rank: 00076
   TM values: Max =   229  Min =  -286  Ave =    -4.51  Std =    26.40
   Min TM value:-286 at an offset of 124.350 usec -> 2021-06-28T17:47:50.577.039.075 (XO: 38.577042052)
   Max TM value:229 at an offset of 126.975 usec -> 2021-06-28T17:47:50.577.041.700 (XO: 38.577044677)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 126.725 usec -> 2021-06-28T17:47:50.577.041.450 (XO: 38.577044427)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 124.163 usec -> 2021-06-28T17:47:50.577.038.887 (XO: 38.577041865)
