-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 08:24:32 2023 UTC
-- File 20210628_111329_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.524206 to 15.673774
-each separately from 11s to 15s and from 524206us to 673774us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.148990 in 1 packets: 18
MIU packet time from 2021-06-28T11:13:40.524.206.000 to 2021-06-28T11:13:44.673.774.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.148990 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9308146 to 16.9308003
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999857 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      BF -> 34.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      5 in 1 packets: 1
      6 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      8 in 1 packets: 19
      9 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      5 in 1 packets: 1
      6 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      8 in 1 packets: 19
      9 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210628_111329_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4550214 to 13.4553104
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.1489750  -- OK for SL --
   MIU time from 2021-06-28T11:13:40.524.206.000 to 2021-06-28T11:13:40.524.495.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.148990

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6042854 to 17.6045744
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-28T11:13:44.673.485.000 to 2021-06-28T11:13:44.673.774.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.455021400 corrected by 40bit 10MHz counter of 12.930814600 obtained at 1Hz counter of 11. Result: 11.524208687
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-28T11:13:40.524.065.725
   Max 1st sample MIU time among all SL packets: 2021-06-28T11:13:44.673.633.725
   Min 1st sample XO time among all SL packets: 11.524068412
   Max 1st sample XO time among all SL packets: 15.673636250
   TM values from -284 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-28T11:13:40.524.065.725 (XO: 11.524068412 - 11.524374413, t=35.6 degC)  Rank: 00090
   TM values: Max =   243  Min =  -278  Ave =    -4.45  Std =    24.58
   Min TM value:-278 at an offset of 124.900 usec -> 2021-06-28T11:13:40.524.190.625 (XO: 11.524193312)
   Max TM value:243 at an offset of 124.625 usec -> 2021-06-28T11:13:40.524.190.350 (XO: 11.524193037)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 127.338 usec -> 2021-06-28T11:13:40.524.193.062 (XO: 11.524195750)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 124.763 usec -> 2021-06-28T11:13:40.524.190.487 (XO: 11.524193175)

 - Block number: 2 starting at 2021-06-28T11:13:44.673.344.725 (XO: 15.673347249 - 15.673653250, t=34.0 degC)  Rank: 00089
   TM values: Max =   232  Min =  -284  Ave =    -4.43  Std =    21.92
   Min TM value:-284 at an offset of 112.625 usec -> 2021-06-28T11:13:44.673.457.350 (XO: 15.673459874)
   Max TM value:232 at an offset of 112.325 usec -> 2021-06-28T11:13:44.673.457.050 (XO: 15.673459574)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 115.013 usec -> 2021-06-28T11:13:44.673.459.737 (XO: 15.673462262)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 112.425 usec -> 2021-06-28T11:13:44.673.457.150 (XO: 15.673459674)
