-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jul 23 05:31:00 2021 UTC
-- File 20210628_061704_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.241202 to 23.409003
-each separately from 3s to 23s and from 241202us to 409003us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.167223 in 1 packets: 18
MIU packet time from 2021-06-28T06:17:07.241.202.000 to 2021-06-28T06:17:27.409.003.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.167223 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8862602 to 24.8861883
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999281 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210628_061704_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.1274629 to 5.1277519
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1671510  -- OK for SL --
   MIU time from 2021-06-28T06:17:07.241.202.000 to 2021-06-28T06:17:07.241.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.167223

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2949029 to 25.2951919
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-28T06:17:27.408.714.000 to 2021-06-28T06:17:27.409.003.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.127462900 corrected by 40bit 10MHz counter of 4.886260200 obtained at 1Hz counter of 3. Result: 3.241203568
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-28T06:17:07.241.061.725
   Max 1st sample MIU time among all SL packets: 2021-06-28T06:17:27.408.862.725
   Min 1st sample XO time among all SL packets: 3.241063293
   Max 1st sample XO time among all SL packets: 23.408864797
   TM values from -313 to 266 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-28T06:17:07.241.061.725 (XO: 3.241063293 - 3.241369294, t=35.6 degC)  Rank: 00079
   TM values: Max =   259  Min =  -313  Ave =    -4.48  Std =    30.06
   Min TM value:-313 at an offset of 126.838 usec -> 2021-06-28T06:17:07.241.188.562 (XO: 3.241190131)
   Max TM value:259 at an offset of 124.263 usec -> 2021-06-28T06:17:07.241.185.987 (XO: 3.241187556)
   Max dTM/dt [1/us]:2.57e+03 at an offset of 126.963 usec -> 2021-06-28T06:17:07.241.188.687 (XO: 3.241190256)
   Min dTM/dt [1/us]:-2.86e+03 at an offset of 124.388 usec -> 2021-06-28T06:17:07.241.186.112 (XO: 3.241187681)

 - Block number: 2 starting at 2021-06-28T06:17:27.408.573.725 (XO: 23.408575796 - 23.408881797, t=35.6 degC)  Rank: 00079
   TM values: Max =   266  Min =  -283  Ave =    -4.50  Std =    33.41
   Min TM value:-283 at an offset of 168.076 usec -> 2021-06-28T06:17:27.408.741.800 (XO: 23.408743871)
   Max TM value:266 at an offset of 113.950 usec -> 2021-06-28T06:17:27.408.687.675 (XO: 23.408689746)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 113.825 usec -> 2021-06-28T06:17:27.408.687.550 (XO: 23.408689621)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 181.538 usec -> 2021-06-28T06:17:27.408.755.263 (XO: 23.408757334)
